Electromagnetic flowmeter features
Frequency programmable low frequency rectangular wave excitation improves stability of flow measurement and low power loss.
The new 16-bit ultra-low power microprocessor with FLASH memory has high integration, fast calculation speed and high calculation accuracy.
Full digital processing, strong anti-interference ability and reliable measurement.
Ultra-low EMI switching power supply, suitable for power supply voltage variation range, high efficiency, small temperature rise; good EMC performance.
Chinese and English menu operation, easy to use, easy to operate, easy to learn and understand.
High definition backlight wide temperature LCD display.
It can perform bidirectional flow measurement and bidirectional total accumulation; it has automatic range switching function, which can effectively improve the measurement accuracy of analog current and frequency output, especially suitable for occasions where the diurnal flow range changes greatly and needs to send control signals; flow measurement range Up to 1500:1.
There are three totalizers inside, which record and display the forward cumulative amount, the reverse cumulative amount and the accumulated difference integrated amount, which are convenient for fluid metering and custody transfer.
Provide isolated or non-isolated RS485/RS232C digital communication interface, and support fieldbus communication modes such as MODBUS, PROFIBUS-DP and HART.
Constant current source fluid resistance measurement can accurately measure the internal resistance of the electrode signal in the case of long-line transmission. It can be used not only to determine whether the fluid in the sensor is empty or not, but also to identify abnormal phenomena such as contamination and coverage of the electrode, and to provide cleaning for the user. Fault processing information such as electrodes.
3. Selection of electrodes and lining
Standard electrodes are used for media that do not produce non-staining electrodes such as crystals, agglomerates, and scales. Otherwise, a doctor blade electrode is used. In addition to the pure alkaline solution, the dosing solution (crystallization of boron and magnesium double salt), the washing liquid and other electromagnetic flowmeters that use the scraper electrode, the other uses the electromagnetic flowmeter of the standard electrode. Since the process pipes in the workshop are basically made of plastic insulation material, if the electromagnetic flowmeter with two electrodes is selected, the grounding ring must be configured to ensure the measurement accuracy. However, the common grounding ring is easy to corrode, and the corrosion-resistant titanium or 钽 grounding ring is high in cost, so the three-electrode electromagnetic flowmeter is selected to ensure measurement accuracy, reduce cost, and convenient installation and maintenance.
4. Flowmeter type selection
The types of electromagnetic flowmeters are divided into: integral type and split type. In Qinghai, the coldest temperature in winter is up to -30 °C. Many metered electromagnetic flowmeters are outdoors, and the environmental humidity of magnesium-lithium separation, pressure filtration and washing processes in lithium and boron systems is large. Exquisite, evaporation, conversion and other processes up to 90 °C. Therefore, the humidity of the high temperature, low temperature, high temperature fluid, vibration source and other working conditions select a split flow meter to effectively prevent the impact on the measuring components, to ensure measurement accuracy and the life of the electromagnetic flowmeter. However, it should be noted that the signal cable between the electromagnetic flowmeter sensor and the transmitter should not be too long, and the manufacturer-specific cable should be used, otherwise the measurement accuracy will be affected. The rest of the working conditions are more ideal.
Pressure and temperature selection
The actual working pressure of the fluid in the pipeline should be less than the rated working pressure of the flowmeter. Pay attention to whether there is negative pressure in the pipeline, such as the evaporation process, and the negative pressure is up to -60KPa. Such working conditions should be communicated with the manufacturer's technicians when ordering. Strict requirements are imposed on the lining material. The actual temperature should meet the temperature requirements specified by the flowmeter.
Power supply selection
The choice of the electromagnetic flowmeter power supply should be selected according to the flowmeter itself and the environment of the site. If the site allows, the power supply is convenient, safe and easy to maintain. As a chemical company, more than 90% of the raw materials and process links are in the liquid phase. During operation, the internal humidity of the workshop is large, and no flammable gas is generated during the whole operation. It is safer to use DC24V for power supply. Outdoor dry environment, choose AC220V power supply, convenient lead wire, good maintenance and low cost.
Some explosion-proof occasions, such as flowmeters used in gas boilers, are ideal for battery-powered electromagnetic flowmeters.