Electromagnetic flowmeter features
Using intelligent judgment, the measurement correction setting is not used, and the air traffic control alarm and electrode detection application are more convenient;
Advanced "rough error handling" technology, which can remove fluids such as slurry to measure sharp disturbances, reduce output runout, maintain high precision measurement and make output more stable;
With a fluid density setting, it can display mass flow;
Constant current excitation current range, 125mA, 250mA optional, can be used with different manufacturers, different types of electromagnetic flow sensors;
Control function with remote reset of the totalizer, with contact signal input for starting and stopping accumulation, suitable for total inspection and batch processing applications;
With self-test and self-diagnosis function;
Advanced non-volatile memory for higher circuit reliability and effective protection of setup and measurement parameters;
The meter can be equipped with an unpowered clock and memory for recording the power down time, power-on time and power-down time;
The meter has an optional hour recording function that can store flow and electrode resistance measurements for more than 30 days.
The new keyboard processing method avoids the keyboard operation affecting the measurement, and can enter and return the operation menu to make the parameter setting more convenient;
The total display uses 10-bit decimal 9999999999 full-value carry, which solves the practice of double-word full value 4294967285 (hexadecimal FFFF) carry-in habit;
With wireless transmission, the mesh network is organized by the wireless HART protocol.
The choice of electromagnetic flowmeter is mainly the correct choice of the transmitter, and the converter only needs to be matched with it.
1. Choice of caliber and range
The diameter of the transmitter is usually the same as that of the piping system. If the piping system is to be designed, the diameter can be selected according to the flow range and flow rate. For electromagnetic flowmeters, the flow rate is suitable for 2 - 4m / s. In special cases, such as solid particles in the liquid, in consideration of wear, the optional flow rate is ≤ 3m / s, for easy to manage the fluid. Available flow rate ≥ 2m / s. After the flow rate is determined, the transmitter diameter can be determined according to qv= D2.
The range of the transmitter can be selected according to two principles: one is that the full scale of the meter is greater than the expected maximum flow value; the other is that the normal flow is greater than 50% of the full scale of the meter to ensure a certain measurement accuracy.
2. Temperature and pressure selection
There are certain restrictions on the fluid pressure and temperature that the electromagnetic flowmeter can measure. When used, the pressure used must be lower than the working pressure specified by the flowmeter. At present, the working pressure specifications of domestically produced electromagnetic flowmeters are:
Less than 50mm caliber, working pressure is 1.6MPa; 900 mm caliber, working pressure is 1 MPa;More than 1000mm caliber, working pressure is 0.6MPa.
If there are special requirements on the pressure resistance of the transmitter, you can negotiate with the manufacturer.
The operating temperature of the electromagnetic flowmeter depends on the lining material used, which is generally 5 - 70 ° C. Such as special treatment, can exceed the above range, such as the wear-resistant corrosion-resistant electromagnetic flowmeter produced by Tianjin Automation Instrument No.3. The transmitter allows the measured medium temperature to be -40 to 130 °C.
3. Selection of lining material and electrode tree material
The lining material and electrode material of the transmitter must be correctly selected according to the physical and chemical properties of the medium. Otherwise, the instrument will be damaged quickly due to the corrosion of the lining and the electrode, and the corrosive medium will easily cause an accident if it leaks. Therefore, the electrode and the lining material must be carefully selected according to the specific measurement medium in the production process.
3. Selection of electrodes and lining
Standard electrodes are used for media that do not produce non-staining electrodes such as crystals, agglomerates, and scales. Otherwise, a doctor blade electrode is used. In addition to the pure alkaline solution, the dosing solution (crystallization of boron and magnesium double salt), the washing liquid and other electromagnetic flowmeters that use the scraper electrode, the other uses the electromagnetic flowmeter of the standard electrode. Since the process pipes in the workshop are basically made of plastic insulation material, if the electromagnetic flowmeter with two electrodes is selected, the grounding ring must be configured to ensure the measurement accuracy. However, the common grounding ring is easy to corrode, and the corrosion-resistant titanium or 钽 grounding ring is high in cost, so the three-electrode electromagnetic flowmeter is selected to ensure measurement accuracy, reduce cost, and convenient installation and maintenance.
4. Flowmeter type selection
The types of electromagnetic flowmeters are divided into: integral type and split type. In Qinghai, the coldest temperature in winter is up to -30 °C. Many metered electromagnetic flowmeters are outdoors, and the environmental humidity of magnesium-lithium separation, pressure filtration and washing processes in lithium and boron systems is large. Exquisite, evaporation, conversion and other processes up to 90 °C. Therefore, the humidity of the high temperature, low temperature, high temperature fluid, vibration source and other working conditions select a split flow meter to effectively prevent the impact on the measuring components, to ensure measurement accuracy and the life of the electromagnetic flowmeter. However, it should be noted that the signal cable between the electromagnetic flowmeter sensor and the transmitter should not be too long, and the manufacturer-specific cable should be used, otherwise the measurement accuracy will be affected. The rest of the working conditions are more ideal.