Electromagnetic flowmeter features
Using intelligent judgment, the measurement correction setting is not used, and the air traffic control alarm and electrode detection application are more convenient;
Advanced "rough error handling" technology, which can remove fluids such as slurry to measure sharp disturbances, reduce output runout, maintain high precision measurement and make output more stable;
With a fluid density setting, it can display mass flow;
Constant current excitation current range, 125mA, 250mA optional, can be used with different manufacturers, different types of electromagnetic flow sensors;
Control function with remote reset of the totalizer, with contact signal input for starting and stopping accumulation, suitable for total inspection and batch processing applications;
With self-test and self-diagnosis function;
Advanced non-volatile memory for higher circuit reliability and effective protection of setup and measurement parameters;
The meter can be equipped with an unpowered clock and memory for recording the power down time, power-on time and power-down time;
The meter has an optional hour recording function that can store flow and electrode resistance measurements for more than 30 days.
The new keyboard processing method avoids the keyboard operation affecting the measurement, and can enter and return the operation menu to make the parameter setting more convenient;
The total display uses 10-bit decimal 9999999999 full-value carry, which solves the practice of double-word full value 4294967285 (hexadecimal FFFF) carry-in habit;
With wireless transmission, the mesh network is organized by the wireless HART protocol.
In order to make the electromagnetic flowmeter work stably and reliably, the following aspects should be noted when selecting the installation location:
1. Try to avoid ferromagnetic objects and equipment with strong electromagnetic fields (large motors, large transformers, etc.) to prevent the magnetic field from affecting the working magnetic field and flow signal of the sensor.
2. It should be installed in a dry and ventilated place to avoid sun and rain. The ambient temperature should be -20~+60°C, and the relative humidity is less than 85%.
3. There should be plenty of space around the flowmeter for easy installation and maintenance.
The measurement principle of the electromagnetic flowmeter does not depend on the characteristics of the flow. If there is a certain turbulence and vortex in the pipeline, it will be generated in the non-measurement zone (such as elbow, tangential current limit or half-opening shut-off valve upstream). Nothing.
If steady-state eddy currents in the measurement zone affect the stability of the measurement and the accuracy of the measurement, then some measures should be taken to stabilize the flow rate distribution:
a.increase the length of the straight pipe before and after; b. use a flow stabilizer; c. reduce the cross section of the measuring point.
The electromagnetic flowmeter has a large measuring range, usually 20:1~50:1, and the optional flow range is wide; the diameter of the electromagnetic flowmeter is wider than other types of flowmeters, from a few millimeters to 3 meters; the positive and negative two-way flow can be measured. The pulsating flow rate can also be measured as long as the pulsation frequency is much lower than the excitation frequency; the meter output is linear in nature; it is easy to select the material type of the fluid contact member, and can be applied to corrosive fluids and the like. Since the electromagnetic flowmeter measures much more with suspended solids or dirt than other flow meters, the probability of failure of the inner wall adhesion layer is relatively high. If the electrical conductivity of the adhesion layer is close to the liquid conductivity, the instrument can output the signal normally, but only change the flow area to form a hidden fault of the measurement error; if it is a high conductivity adhesion layer, the electromotive force between the electrodes will be short-circuited; if it is an insulating adhesion layer The surface of the electrode is insulated to disconnect the measuring circuit. The latter two phenomena will make the meter inoperable.