Plug-in ultrasonic flowmeter features
Programmable frequency low frequency rectangular wave excitation improves stability of flow measurement and low power loss;
Using 16-bit embedded microprocessor, the operation speed is fast and the precision is high;
Full digital processing, strong anti-interference ability, reliable measurement, high precision, flow measurement range up to 100:1;
Ultra-low EMI switching power supply, suitable for wide range of power supply voltage and good anti-EMC performance;
Full Chinese character menu operation, easy to use, easy to operate, easy to learn and understand;
High definition backlit LCD display;
It has two-way flow measurement and two-way total accumulation function, and the current and frequency have bidirectional output function;
There are three totalizers in the internal to display the forward cumulative amount, the reverse cumulative amount and the difference integrated amount;
With RS485 or RS232 digital communication signal output
It has a conductivity measurement function to determine whether the sensor is empty or not;
The constant current excitation current range is large, and can be used with different companies and different types of electromagnetic flow sensors;
With self-test and self-diagnosis function;
High reliability with SMD devices and surface mount (SMT) technology;
The internal design of the meter has a power-down clock that can record 16 power-down times.
In order to make the electromagnetic flowmeter work stably and reliably, the following aspects should be noted when selecting the installation location:
1. Try to avoid ferromagnetic objects and equipment with strong electromagnetic fields (large motors, large transformers, etc.) to prevent the magnetic field from affecting the working magnetic field and flow signal of the sensor.
2. It should be installed in a dry and ventilated place to avoid sun and rain. The ambient temperature should be -20~+60°C, and the relative humidity is less than 85%.
3. There should be plenty of space around the flowmeter for easy installation and maintenance.
The measurement principle of the electromagnetic flowmeter does not depend on the characteristics of the flow. If there is a certain turbulence and vortex in the pipeline, it will be generated in the non-measurement zone (such as elbow, tangential current limit or half-opening shut-off valve upstream). Nothing.
If steady-state eddy currents in the measurement zone affect the stability of the measurement and the accuracy of the measurement, then some measures should be taken to stabilize the flow rate distribution:
a.increase the length of the straight pipe before and after; b. use a flow stabilizer; c. reduce the cross section of the measuring point.
Accuracy class and function According to the measurement requirements and the use occasions, the instrument accuracy level is selected to achieve economical efficiency. For example, in the case of trade settlement, product handover and energy measurement, the accuracy level should be higher, such as 1.0, 0.5, or higher; for process control, select different accuracy levels according to control requirements; It is to detect the process flow, no need to do precise control and measurement, you can choose a lower accuracy level, such as 1.5, 2.5, or even 4.0, then you can use a low-cost plug-in electromagnetic flowmeter.
Measuring medium flow rate, meter range and diameter When measuring general medium, the full flow of electromagnetic flowmeter can be selected within the range of 0.5-12m/s of measuring medium flow, and the range is wider. The meter specification (caliber) is not necessarily the same as the process pipeline. It should be determined whether the measured flow range is within the flow rate range. That is, when the pipeline flow rate is too low to meet the flow meter requirements or the measurement accuracy cannot be guaranteed at this flow rate, It is necessary to reduce the gauge diameter, thereby increasing the flow rate inside the tube and obtaining satisfactory measurement results.
Try to avoid ferromagnetic objects and equipment with strong electromagnetic fields to prevent the magnetic field from affecting the working magnetic field and flow signal of the sensor.
Should be installed in the dry and ventilated place, to avoid sun and rain, the ambient temperature should be -20 ~ +60 ° C, relative humidity is less than 85%.
There should be ample space around the flowmeter for easy testing and maintenance.