Because all kinds of ultrasonic flowmeters can be installed outside the pipe, non-contact flow measurement, the cost of the instrument is basically independent of the size of the pipe to be tested, while other types of flowmeters increase with the increase in caliber, so the cost is increased. The flowmeter is superior to the other functions of the same type of flowmeter. It is considered to be a good large-diameter flow measuring instrument. The Doppler ultrasonic flowmeter can measure the flow of two-phase medium, so it can be used for the measurement of dirty sewage such as sewers and sewage. In power plants, the use of portable ultrasonic flowmeters to measure large pipe diameters such as turbine water inflow and turbine circulating water is much more convenient than in the past. Ultrasonic flow juice can also be used for gas measurement. Pipe diameters range from 2cm to 5m, from a few meters wide open channels, culverts to 500m wide rivers.
In addition, the accuracy of the flow measurement of the ultrasonic measuring instrument is almost independent of the temperature, pressure, viscosity, density and other parameters of the measured fluid, and can be made into non-contact and portable measuring instruments, so it can solve the problem that other types of instruments are difficult to measure. Flow measurement problems for corrosive, non-conductive, radioactive, and flammable and explosive media. In addition, in view of the non-contact measurement characteristics, coupled with reasonable electronic circuits, one instrument can adapt to a variety of pipe diameter measurements and a variety of flow range measurements. The adaptability of ultrasonic flowmeters is also unmatched by other instruments. Ultrasonic flowmeters have some of the above advantages, so it has received more and more attention and has been developed into a series of products and generalization. It has been made into standard, high-temperature, explosion-proof and wet instruments of different channels to adapt to different media. Flow measurement for occasions and different pipeline conditions.
In order to make the electromagnetic flowmeter work stably and reliably, the following aspects should be noted when selecting the installation location:
1. Try to avoid ferromagnetic objects and equipment with strong electromagnetic fields (large motors, large transformers, etc.) to prevent the magnetic field from affecting the working magnetic field and flow signal of the sensor.
2. It should be installed in a dry and ventilated place to avoid sun and rain. The ambient temperature should be -20~+60°C, and the relative humidity is less than 85%.
3. There should be plenty of space around the flowmeter for easy installation and maintenance.
The measurement principle of the electromagnetic flowmeter does not depend on the characteristics of the flow. If there is a certain turbulence and vortex in the pipeline, it will be generated in the non-measurement zone (such as elbow, tangential current limit or half-opening shut-off valve upstream). Nothing.
If steady-state eddy currents in the measurement zone affect the stability of the measurement and the accuracy of the measurement, then some measures should be taken to stabilize the flow rate distribution:
a.increase the length of the straight pipe before and after; b. use a flow stabilizer; c. reduce the cross section of the measuring point.
1. Requirements for straight pipe sections
Inlet/outlet straight pipe section: inlet should be ≥10×DN; outlet should be ≥5×DN
2. Grounding point requirement
In order to make the instrument work reliably, improve the measurement accuracy, free from external parasitic potential / the sensor should have good grounding, the grounding resistance is less than 10. (If the metal pipe is well grounded, no special grounding device is required) 3.3 Pair installation Location requirements as shown
Insert the electromagnetic flowmeter according to the situation of the pipeline in the field. If the flowmeter without the ball valve is installed, it should be installed on the pipeline without overpressure (that is, the flowmeter without the ball valve can be installed without pressure), and the hole is opened in the pipeline. Diameter 50, ready to weld the connecting welded pipe to the opening of the pipe; for occasions requiring constant flow loading or unloading or no media spillage, a ball valve must be installed, that is, a plug-in electromagnetic flowmeter with a ball valve structure is selected; With a diameter of 50, it is ready to weld the connecting welded pipe to the opening of the pipe.
Measuring range : Recommended range: 0.5m/s to 10m/s continuously adjustable. Maximum use range: continuously adjustable from 0.2m/s to 15m/s
Signal output: 1, the switch quantity can be set to: pulse output (up to 1000HZ); high/low flow alarm; empty pipe alarm; flow direction indication;
Fault alarm; 2, current output: 4-20mA output
Configuration method: 1. Field configuration through three manual keys. 2. Field configuration via remote control. 3. Perform on-site configuration through the handheld communicator.
Memory: The memory that the EEPROM does not disappear, no battery saving.