Electrode: Its function is to extract and induce a proportional induced inductive potential signal. The electrodes are typically made of non-magnetically conductive stainless steel and are required to be flush with the liner so that the fluid passes unimpeded. It should be installed in the vertical direction of the pipe to prevent deposits from accumulating on it and affecting the measurement accuracy.
Enclosure: Made of ferromagnetic material, it is the cover of the distribution system excitation coil and isolates the interference of external magnetic field.
Lining: A complete electrical insulation lining on the inside of the measuring tube and on the flange sealing surface. It directly contacts the liquid to be measured, and its function is to increase the corrosion resistance of the measuring catheter and prevent the induced potential from being short-circuited by the metal measuring tube wall. Most of the lining materials are PTFE plastics and ceramics that are resistant to corrosion, high temperature and wear.
Converter: The induced potential signal generated by the liquid flow is very weak and is greatly affected by various interference factors. The function of the converter is to amplify and convert the induced potential signal into a unified standard signal and suppress the main interference signal. Its task is to amplify the induced potential signal Ex detected by the electrode into a unified standard DC signal.
Application of electromagnetic flowmeter
Because electromagnetic flowmeter has its unique advantages, it is widely used in chemical, chemical fiber, food, paper, sugar, water supply and drainage, environmental protection, water conservancy, steel, petroleum, pharmaceutical and other industrial fields. Various acid and alkali salt solutions, mud, pulp, pulp, coal water slurry, corn syrup, fiber pulp, lime slurry, water supply and drainage, brine, hydrogen peroxide, beer, wort, various beverages, black liquor, green liquor, etc. The volumetric flow of the medium.
The main technical parameters
Nominal diameter series DN (mm)
Pipeline PTFE lining:
Pipeline rubber lining:
Special specifications can be customized
Flow direction: current output
Positive, negative, net flow
Range ratio: 150:1
Repeatability error: ±0.1 of the measured value
Accuracy level: pipeline type: 0.5 level, 1.0 level
Measured medium temperature
Ordinary rubber lining: -20 ~ +60 ° C
High temperature rubber lining: -20 ~ +90 ° C
PTFE lining: -30 ~ +100 ° C
High temperature PTFE lining: -30 ~ +180 ° C
Rated operating voltage
Pipe type: DN6-DN100≤1.6Mpa, DN125-DN225≤1.0Mpa high pressure can be customized
Conductivity of the measured fluid ≥ 5us / cm (integrated)
Most water-based media have a conductivity of 200-800 us/cm, and electromagnetic flow meters can be used to measure the flow rate.
Load resistance: 0 ~ 10mA, 0 ~ 1.5kΩ
4 to 20 mA, 0 to 750 Ω
Digital frequency output
The upper limit of the output frequency can be set to open the bidirectional output of the open collector of the transistor with optical isolation from 1 to 5000HZ. External ≤35V, the maximum current of the collector is 25mA when conducting
Straight pipe length
Pipeline type: upstream ≥5DN, downstream ≥2DN
The flowmeter and the pipe are flanged, and the flange connection size should meet the requirements of GB11988.
Explosion-proof mark: mdllBT4
Ambient temperature: -10 ° C ~ +55 ° C
Relative temperature: 5% to 90%
Total power consumption: less than 15W
Traditional mechanical flowmeters, such as differential pressure flowmeters, positive displacement flowmeters, and variable area flowmeters, are already in the stage of popularization, with fierce price competition, decreasing profit margins, less technological innovation, and relatively mature markets. . Frost & Sullivan believes that achieving product differentiation and customized production is an important breakthrough for manufacturers in the fierce competition in mature markets. Based on Frost & Sullivan's analysis of the needs of industry users, the user community expects manufacturers to provide automated equipment that will bring tangible benefits to the production process. Users will have specific requirements in the application process, such as: application in the special environment of the petrochemical industry, the need for rugged design and explosion-proof certification; user demand for Coriolis flowmeter designed for straight pipe. How to effectively obtain the actual needs of users and improve the traditional products is a challenge to the manufacturer's differentiated and customized production process.
Guiding users to accept and use new technology flow meters, such as ultrasonic flow meters, electromagnetic flow meters, and thermal mass flow meters, is another challenge for manufacturers to make the market bigger and stronger. In fact, the new technology flowmeter mentioned above has been developed and applied more than ten years ago. How to make customers realize that using the new technology flowmeter can effectively improve the production efficiency is an important issue for manufacturers.
In addition, new technology flowmeters are constantly being introduced into various industries, and fast and efficient after-sales service is also crucial for manufacturers. In particular, the use of Flowfield based on Foundation Fieldbus and Profibus PA bus has certain requirements for software technology. Effective services can provide users with more suitable solutions and close to users.