Ultrasonic flowmeter uses the electronic principle to measure the flow meter. Although it has its own verification procedure, because it has the same usage management mode as the tap water industry measurement, it puts its performance requirements, measurement accuracy, verification cycle, etc. into the cold water meter. In the meantime, it has become more standardized. Since then, the use of large-caliber ultrasonic flowmeters has normative documents in the water industry.
Ultrasonic flowmeter (ultrasonic water meter) Because the measurement principle is different from electromagnetic water meter, its accuracy is different. How to choose an ultrasonic flowmeter with superior performance and precision:
The user flow range is used to determine the type, diameter, temperature, medium, etc. of the ultrasonic flowmeter, especially considering the common flow rate and the starting flow rate. The purpose is to ensure the water supply quantity, accurately measure and reduce the leakage rate. When using the water meter, the user's water meter Commonly used traffic generally cannot exceed the common flow value specified by the product.
Different flow rate to determine the diameter of the flow meter can meet the measurement range.
Different user environments.
Also consider the working pressure, the structure of the flow certificate, the form of the converter (whether bidirectional flow, current, pulse, communication interface, data storage) and the length of the connecting cable.
The performance and accuracy of the ultrasonic flowmeter are selected.
Users can learn more about the peers who are already using the ultrasonic flowmeter, to see the effect of their use, the accuracy of the measurement.
At present, industrial flow measurement generally has problems of large diameter, large flow, small diameter, small flow, high temperature liquid, and gas measurement. This is because the general flowmeter will bring manufacturing and transportation as the diameter of the measuring pipe increases. The difficulty of the above, the advantages of increased cost, increased energy loss, and inconvenient installation, ultrasonic flowmeters can be avoided. The German FLEXIM ultrasonic flowmeter measures diameters from 6mm to 6500 mm, from a few meters wide open channels, culverts to 500m wide rivers.
Application of electromagnetic flowmeter
Because electromagnetic flowmeter has its unique advantages, it is widely used in chemical, chemical fiber, food, paper, sugar, water supply and drainage, environmental protection, water conservancy, steel, petroleum, pharmaceutical and other industrial fields. Various acid and alkali salt solutions, mud, pulp, pulp, coal water slurry, corn syrup, fiber pulp, lime slurry, water supply and drainage, brine, hydrogen peroxide, beer, wort, various beverages, black liquor, green liquor, etc. The volumetric flow of the medium.
The main technical parameters
Nominal diameter series DN (mm)
Pipeline PTFE lining:
Pipeline rubber lining:
Special specifications can be customized
Flow direction: current output
Positive, negative, net flow
Range ratio: 150:1
Repeatability error: ±0.1 of the measured value
Accuracy level: pipeline type: 0.5 level, 1.0 level
Measured medium temperature
Ordinary rubber lining: -20 ~ +60 ° C
High temperature rubber lining: -20 ~ +90 ° C
PTFE lining: -30 ~ +100 ° C
High temperature PTFE lining: -30 ~ +180 ° C
Rated operating voltage
Pipe type: DN6-DN100≤1.6Mpa, DN125-DN225≤1.0Mpa high pressure can be customized
Conductivity of the measured fluid ≥ 5us / cm (integrated)
Most water-based media have a conductivity of 200-800 us/cm, and electromagnetic flow meters can be used to measure the flow rate.
Load resistance: 0 ~ 10mA, 0 ~ 1.5kΩ
4 to 20 mA, 0 to 750 Ω
Digital frequency output
The upper limit of the output frequency can be set to open the bidirectional output of the open collector of the transistor with optical isolation from 1 to 5000HZ. External ≤35V, the maximum current of the collector is 25mA when conducting
Straight pipe length
Pipeline type: upstream ≥5DN, downstream ≥2DN
The flowmeter and the pipe are flanged, and the flange connection size should meet the requirements of GB11988.
Explosion-proof mark: mdllBT4
Ambient temperature: -10 ° C ~ +55 ° C
Relative temperature: 5% to 90%
Total power consumption: less than 15W
Measuring principle of electromagnetic flowmeter
The principle of electromagnetic flowmeter measurement is based on Faraday's law of electromagnetic induction. The measuring tube of the flow meter is a non-magnetic alloy short tube lined with an insulating material. The two electrodes are fixed to the measuring tube through the tube wall in the tube diameter direction. The electrode tip is substantially flush with the inner surface of the liner. When the exciting coil is excited by the bidirectional square wave pulse, a working magnetic field having a magnetic flux density B is generated in a direction perpendicular to the axis of the measuring tube. At this time, if the fluid having a certain conductivity passes through the measuring tube, the cutting magnetic line induces the electromotive force E. The electromotive force E is proportional to the product of the magnetic flux density B, the inner diameter D of the measuring tube and the average velocity V. The electromotive force E (flow signal) is detected by the electrode and sent to the converter through the cable. After the converter amplifies the flow signal, it can display the fluid flow, and can output signals such as pulse and analog current for flow control and regulation.