The mediums that need to be measured in the site are different. Now, a brief introduction is made on the selection of flow meters for different media.
The selection of large water flow meter: suitable for electromagnetic flowmeter, ultrasonic flowmeter. Other applications such as vortex flowmeters, orifice flowmeters, etc. have faded out of this application, and these two types of flow meters are the most important in the industry. In particular, it should be mentioned that the application of Kaifeng Shengda's plug-in flowmeter and electromagnetic flowmeter are better.
Sewage, pulp and other turbid liquid meter selection: electromagnetic flowmeter can be used. However, when choosing an electromagnetic flowmeter, consider that the liquid does not contain more air or bubbles.
The meter with more bubbles of liquid selection: Doppler ultrasonic flowmeter can be selected, using this type of flowmeter to measure the fluid with bubbles, the effect is very good.
Flow meter selection with low conductivity such as pure water and demineralized water: Turbine flow meter is very suitable for measuring such fluids.
Flow meter selection for strong corrosive media such as acid and lye: 1. Electromagnetic flowmeter with acid and alkali lining. 2. clamp type ultrasonic flowmeter.
Selection of flowmeters for mediums with large concentration and large solid particle content such as mortar and electric powder slurry: electromagnetic flowmeter.
Flow meter selection for oil media such as oil and diesel: turbine flow meter and ultrasonic flow meter.
Gas flow meter selection: vortex flowmeter.
The lining material should be selected according to the corrosiveness, wear and temperature of the tested medium:
First, natural rubber (soft rubber)
1, better elasticity, wear resistance and tearing force
2, resistant to general weak acid, weak alkali corrosion
3. Water and sewage
Second, acid-resistant rubber (hard rubber)
It can withstand the corrosion of hydrochloric acid, acetic acid, oxalic acid, ammonia water, phosphoric acid and 50% sulfuric acid, sodium hydroxide and potassium hydroxide at normal temperature, but it is not resistant to corrosion by strong oxidants.
It can measure general acid, alkali and salt solutions.
Third, neoprene (Neoprene)
1, excellent elasticity, high tearing force, good wear resistance
2, resistant to general low concentration of acid and alkali, salt solution corrosion, but not resistant to oxidation of the medium, and temperature requirements <80 ° C;
It can measure water, sewage, mud and pulp.
Fourth, Polyurethane (Polyurethane)
1, excellent wear resistance
2, poor acid and alkali resistance, temperature requirements <40 ° C
Measure medium-strongly worn coal slurry, mud and slurry
Five, polytetrafluoroethylene (PTFE)
1. Boiling hydrochloric acid, sulfuric acid, nitric acid, aqua regia, concentrated alkali and various organic solvents
2, good wear resistance, poor bonding performance, temperature requirements -80 ~ +180 ° C;
Measuring concentration, concentrated alkali strong corrosive solution and sanitary medium
Ultrasonic flowmeter features
The unique signal digitization processing technology makes the meter measurement signal more stable, anti-interference ability and more accurate measurement.
No mechanical transmission parts are not easily damaged, maintenance-free and have a long service life.
The circuit is more optimized, with high integration, low power consumption and high reliability.
Intelligent standard signal output, friendly man-machine interface, multiple secondary signal output, for you to choose.
Pipe-section small pipe diameter measurement is economical and convenient, and the measurement accuracy is high.
Detailed installation of ultrasonic flowmeter
Ultrasonic flowmeters should be aware of the site prior to installation, including:
1. What is the distance from the host at the installation of the sensor;
2. Pipe material, pipe wall thickness and pipe diameter;
3. Years of pipeline;
4. The type of fluid, whether it contains impurities, bubbles and whether it is full;
5. Fluid temperature;
6. Whether there is interference source at the installation site (such as frequency conversion, strong magnetic field, etc.);
7. Four seasons temperature at the host place;
8. Whether the power supply voltage used is stable;
9. Do you need remote signals and types;
According to the site conditions provided above, the manufacturer can configure the site conditions and, if necessary, special models.