Electrode: Its function is to extract and induce a proportional induced inductive potential signal. The electrodes are typically made of non-magnetically conductive stainless steel and are required to be flush with the liner so that the fluid passes unimpeded. It should be installed in the vertical direction of the pipe to prevent deposits from accumulating on it and affecting the measurement accuracy.
Enclosure: Made of ferromagnetic material, it is the cover of the distribution system excitation coil and isolates the interference of external magnetic field.
Lining: A complete electrical insulation lining on the inside of the measuring tube and on the flange sealing surface. It directly contacts the liquid to be measured, and its function is to increase the corrosion resistance of the measuring catheter and prevent the induced potential from being short-circuited by the metal measuring tube wall. Most of the lining materials are PTFE plastics and ceramics that are resistant to corrosion, high temperature and wear.
Converter: The induced potential signal generated by the liquid flow is very weak and is greatly affected by various interference factors. The function of the converter is to amplify and convert the induced potential signal into a unified standard signal and suppress the main interference signal. Its task is to amplify the induced potential signal Ex detected by the electrode into a unified standard DC signal.
Ultrasonic flowmeter features
The unique signal digitization processing technology makes the meter measurement signal more stable, anti-interference ability and more accurate measurement.
No mechanical transmission parts are not easily damaged, maintenance-free and have a long service life.
The circuit is more optimized, with high integration, low power consumption and high reliability.
Intelligent standard signal output, friendly man-machine interface, multiple secondary signal output, for you to choose.
Pipe-section small pipe diameter measurement is economical and convenient, and the measurement accuracy is high.
Detailed installation of ultrasonic flowmeter
Ultrasonic flowmeters should be aware of the site prior to installation, including:
1. What is the distance from the host at the installation of the sensor;
2. Pipe material, pipe wall thickness and pipe diameter;
3. Years of pipeline;
4. The type of fluid, whether it contains impurities, bubbles and whether it is full;
5. Fluid temperature;
6. Whether there is interference source at the installation site (such as frequency conversion, strong magnetic field, etc.);
7. Four seasons temperature at the host place;
8. Whether the power supply voltage used is stable;
9. Do you need remote signals and types;
According to the site conditions provided above, the manufacturer can configure the site conditions and, if necessary, special models.
Ultrasonic flowmeter classification
Plug-in ultrasonic flowmeter: can be installed and maintained without interruption. Ceramic sensors are used for non-stop production installation using dedicated drilling equipment. Generally for mono measurement, in order to improve measurement accuracy, three channels can be selected.
Pipe-type ultrasonic flowmeter: the pipeline installation needs to be cut, but the subsequent maintenance can be stopped. Mono or 3-channel sensors are available.
External clip-on ultrasonic flowmeter: capable of performing fixed and mobile measurements. It is installed with a special coupling agent (silicone rubber cured at room temperature or high-temperature long-chain polymer grease) and does not damage the pipeline during installation.
Portable Ultrasonic Flowmeter: Portable, built-in rechargeable lithium battery, suitable for mobile measurement, with magnetic sensor.
1, non-contact measurement method, small size, easy to carry
2, suitable for on-site measurement of various sizes of pipe sound guiding media
3, built-in nickel-metal hydride rechargeable battery working time of more than 20 hours
4, user interface is flexible, easy to use
5, intelligent on-site printing function to ensure the integrity of the flow data
6, equipped with an integrated aluminum alloy protective box, can be used in harsh outdoor environments
Hand-held ultrasonic flowmeter: small size, light weight, built-in rechargeable lithium battery, hand-held, with magnetic sensor.
Explosion-proof ultrasonic flowmeter: used for explosive liquid flow measurement, it is explosion-proof and intrinsically safe. That is, the converter is explosion-proof and the sensor is intrinsically safe.