The choice of electromagnetic flowmeter is mainly the correct choice of the transmitter, and the converter only needs to be matched with it.
Choice of caliber and range
The diameter of the transmitter is usually the same as that of the piping system. If the piping system is to be designed, the diameter can be selected according to the flow range and flow rate. For electromagnetic flowmeters, the flow rate is suitable for 2 - 4m / s. In special cases, such as solid particles in the liquid, in consideration of wear, the optional flow rate is ≤ 3m / s, for easy to manage the fluid. Available flow rate ≥ 2m / s. After the flow rate is determined, the transmitter diameter can be determined according to qv= D2.
The range of the transmitter can be selected according to two principles: one is that the full scale of the meter is greater than the expected maximum flow value; the other is that the normal flow is greater than 50% of the full scale of the meter to ensure a certain measurement accuracy.
Temperature and pressure selection
There are certain restrictions on the fluid pressure and temperature that the electromagnetic flowmeter can measure. When used, the pressure used must be lower than the working pressure specified by the flowmeter. At present, the working pressure specifications of domestically produced electromagnetic flowmeters are:
Less than 50mm caliber, working pressure is 1.6MPa; 900 mm caliber, working pressure is 1 MPa;More than 1000mm caliber, working pressure is 0.6MPa.
If there are special requirements on the pressure resistance of the transmitter, you can negotiate with the manufacturer.
The operating temperature of the electromagnetic flowmeter depends on the lining material used, which is generally 5 - 70 ° C. Such as special treatment, can exceed the above range, such as the wear-resistant corrosion-resistant electromagnetic flowmeter produced by Tianjin Automation Instrument No.3. The transmitter allows the measured medium temperature to be -40 to 130 °C.
Selection of lining material and electrode tree material
The lining material and electrode material of the transmitter must be correctly selected according to the physical and chemical properties of the medium. Otherwise, the instrument will be damaged quickly due to the corrosion of the lining and the electrode, and the corrosive medium will easily cause an accident if it leaks. Therefore, the electrode and the lining material must be carefully selected according to the specific measurement medium in the production process.
The lining material should be selected according to the corrosiveness, wear and temperature of the tested medium:
First, natural rubber (soft rubber)
1, better elasticity, wear resistance and tearing force
2, resistant to general weak acid, weak alkali corrosion
3. Water and sewage
Second, acid-resistant rubber (hard rubber)
It can withstand the corrosion of hydrochloric acid, acetic acid, oxalic acid, ammonia water, phosphoric acid and 50% sulfuric acid, sodium hydroxide and potassium hydroxide at normal temperature, but it is not resistant to corrosion by strong oxidants.
It can measure general acid, alkali and salt solutions.
Third, neoprene (Neoprene)
1, excellent elasticity, high tearing force, good wear resistance
2, resistant to general low concentration of acid and alkali, salt solution corrosion, but not resistant to oxidation of the medium, and temperature requirements <80 ° C;
It can measure water, sewage, mud and pulp.
Fourth, Polyurethane (Polyurethane)
1, excellent wear resistance
2, poor acid and alkali resistance, temperature requirements <40 ° C
Measure medium-strongly worn coal slurry, mud and slurry
Five, polytetrafluoroethylene (PTFE)
1. Boiling hydrochloric acid, sulfuric acid, nitric acid, aqua regia, concentrated alkali and various organic solvents
2, good wear resistance, poor bonding performance, temperature requirements -80 ~ +180 ° C;
Measuring concentration, concentrated alkali strong corrosive solution and sanitary medium
Ultrasonic flowmeter classification
Plug-in ultrasonic flowmeter: can be installed and maintained without interruption. Ceramic sensors are used for non-stop production installation using dedicated drilling equipment. Generally for mono measurement, in order to improve measurement accuracy, three channels can be selected.
Pipe-type ultrasonic flowmeter: the pipeline installation needs to be cut, but the subsequent maintenance can be stopped. Mono or 3-channel sensors are available.
External clip-on ultrasonic flowmeter: capable of performing fixed and mobile measurements. It is installed with a special coupling agent (silicone rubber cured at room temperature or high-temperature long-chain polymer grease) and does not damage the pipeline during installation.
Portable Ultrasonic Flowmeter: Portable, built-in rechargeable lithium battery, suitable for mobile measurement, with magnetic sensor.
1, non-contact measurement method, small size, easy to carry
2, suitable for on-site measurement of various sizes of pipe sound guiding media
3, built-in nickel-metal hydride rechargeable battery working time of more than 20 hours
4, user interface is flexible, easy to use
5, intelligent on-site printing function to ensure the integrity of the flow data
6, equipped with an integrated aluminum alloy protective box, can be used in harsh outdoor environments
Hand-held ultrasonic flowmeter: small size, light weight, built-in rechargeable lithium battery, hand-held, with magnetic sensor.
Explosion-proof ultrasonic flowmeter: used for explosive liquid flow measurement, it is explosion-proof and intrinsically safe. That is, the converter is explosion-proof and the sensor is intrinsically safe.