To ensure the measurement accuracy of the electromagnetic flowmeter, proper installation is very important.
The transmitter should be installed in a dry and ventilated place indoors. Avoid installation in places where the ambient temperature is too high, should not be subject to strong vibration, try to avoid equipment with strong magnetic fields, such as large motors, transformers, etc. Avoid installation in corrosive gases. The installation location is easy to overhaul. This is the environmental condition to ensure the normal operation of the transmitter.
In order to ensure that the measuring tube of the transmitter is filled with the measured medium, the stacker is preferably installed vertically, and the flow direction is from bottom to top. Especially for liquid-solid two-phase flow, it must be installed vertically. If the site is only allowed to be installed horizontally, it must be ensured that the two electrodes are at the same level.
The transmitter should be equipped with valves and bypass at both ends.
The millivolt AC potential measured by the electrode of the electromagnetic flow transmitter is based on the liquid potential in the transmitter. In order to stabilize the liquid potential and maintain the potential of the transmitter and fluid to ensure stable measurement, the transmitter shell and the metal tube should have good grounding at both ends, and the converter housing should also be grounded. The grounding resistance should not be greater than 10 and cannot be shared with the grounding wire of other electrical equipment. If the transmitter housing is not guaranteed to be in good contact with the metal pipe, connect them with metal wires. Re-reliable grounding.
In order to avoid interference signals, the signal between the transmitter and the converter must be transmitted with shielded wires. It is not allowed to place the signal cable and power cable in parallel in the same cable. The length of the signal cable should generally not exceed 30 m.
Converter installation site should avoid AC and DC strong magnetic field and vibration, ambient temperature is -20 to 50 ° C, does not contain corrosive gases, relative humidity is not more than 80%.
In order to avoid the influence of the flow rate on the relative measurement, the flow regulating valve should be placed downstream of the transmitter. For small-caliber transmitters, since the distance from the center of the electrode to the inlet end of the flowmeter is equivalent to several times the length of the diameter D, the upstream straight pipe may not be specified. However, for a flowmeter with a large diameter, generally there should be a straight pipe section of 5D or more in the upstream, and the downstream pipe section is generally not required.
The lining material should be selected according to the corrosiveness, wear and temperature of the tested medium:
First, natural rubber (soft rubber)
1, better elasticity, wear resistance and tearing force
2, resistant to general weak acid, weak alkali corrosion
3. Water and sewage
Second, acid-resistant rubber (hard rubber)
It can withstand the corrosion of hydrochloric acid, acetic acid, oxalic acid, ammonia water, phosphoric acid and 50% sulfuric acid, sodium hydroxide and potassium hydroxide at normal temperature, but it is not resistant to corrosion by strong oxidants.
It can measure general acid, alkali and salt solutions.
Third, neoprene (Neoprene)
1, excellent elasticity, high tearing force, good wear resistance
2, resistant to general low concentration of acid and alkali, salt solution corrosion, but not resistant to oxidation of the medium, and temperature requirements <80 ° C;
It can measure water, sewage, mud and pulp.
Fourth, Polyurethane (Polyurethane)
1, excellent wear resistance
2, poor acid and alkali resistance, temperature requirements <40 ° C
Measure medium-strongly worn coal slurry, mud and slurry
Five, polytetrafluoroethylene (PTFE)
1. Boiling hydrochloric acid, sulfuric acid, nitric acid, aqua regia, concentrated alkali and various organic solvents
2, good wear resistance, poor bonding performance, temperature requirements -80 ~ +180 ° C;
Measuring concentration, concentrated alkali strong corrosive solution and sanitary medium
Use and promotion of electromagnetic flowmeter
The intelligent electromagnetic flowmeter is a fully intelligent electromagnetic flowmeter developed by our company using advanced technology at home and abroad. Its all-Chinese electromagnetic converter core adopts high-speed central processing unit, which has fast calculation speed, high precision and reliable measurement performance. The converter circuit design adopts the latest international technology, the input impedance is up to 1015 ohms, the common mode rejection ratio is better than 100db, and the external interference and 60Hz/50Hz interference suppression ability is better than 90db, which can measure the lower conductivity fluid medium flow. The sensor adopts a new non-uniform magnetic field technology and a special magnetic circuit structure. The magnetic field is stable and reliable, and the volume is greatly reduced, the weight is reduced, and the flowmeter has the characteristics of small size and light weight. Adhering to the enterprise spirit of "seeking truth, hard work, innovation and development", we insist on "customer first, technology first, high quality, first-class service", we will provide you with first-class products and first-class service.