Instruments that measure fluid flow are collectively referred to as flow meters or flow meters. The flowmeter is one of the important instruments in industrial measurement. With the development of industrial production, the accuracy and range of flow measurement requirements are getting higher and higher, and the flow measurement technology is changing with each passing day. Various types of flow meters have been introduced to suit various applications. More than 100 flow meters have been put into use. From different perspectives, flow meters have different classification methods. There are two commonly used classification methods. One is to classify according to the measurement principle adopted by the flowmeter: the second is to classify according to the structural principle of the flowmeter.
Sort by measurement principle
a. Mechanical principle:
Instruments belonging to such principles have differential pressure type, rotor type using Bernoulli's theorem;
Impulse type using the momentum theorem, movable tube type;
Direct mass equation using Newton's second law;
a target using the principle of fluid momentum;
Turbine using the angular momentum theorem;
Vortex type using vortex principle of fluid oscillation;
Use the total static pressure difference of the pitot tube type as well as volumetric and sputum, trough and so on.
b. Electrical principle:
Differential capacitor type,
Strain resistance type, etc.
c. Acoustic principle:
d. Thermal principles:
Indirect calorimetry and so on.
1. Requirements for straight pipe sections
Inlet/outlet straight pipe section: inlet should be ≥10×DN; outlet should be ≥5×DN
2. Grounding point requirement
In order to make the instrument work reliably, improve the measurement accuracy, free from external parasitic potential / the sensor should have good grounding, the grounding resistance is less than 10. (If the metal pipe is well grounded, no special grounding device is required) 3.3 Pair installation Location requirements as shown
Insert the electromagnetic flowmeter according to the situation of the pipeline in the field. If the flowmeter without the ball valve is installed, it should be installed on the pipeline without overpressure (that is, the flowmeter without the ball valve can be installed without pressure), and the hole is opened in the pipeline. Diameter 50, ready to weld the connecting welded pipe to the opening of the pipe; for occasions requiring constant flow loading or unloading or no media spillage, a ball valve must be installed, that is, a plug-in electromagnetic flowmeter with a ball valve structure is selected; With a diameter of 50, it is ready to weld the connecting welded pipe to the opening of the pipe.
Measuring range : Recommended range: 0.5m/s to 10m/s continuously adjustable. Maximum use range: continuously adjustable from 0.2m/s to 15m/s
Signal output: 1, the switch quantity can be set to: pulse output (up to 1000HZ); high/low flow alarm; empty pipe alarm; flow direction indication;
Fault alarm; 2, current output: 4-20mA output
Configuration method: 1. Field configuration through three manual keys. 2. Field configuration via remote control. 3. Perform on-site configuration through the handheld communicator.
Memory: The memory that the EEPROM does not disappear, no battery saving.
Use and promotion of electromagnetic flowmeter
The intelligent electromagnetic flowmeter is a fully intelligent electromagnetic flowmeter developed by our company using advanced technology at home and abroad. Its all-Chinese electromagnetic converter core adopts high-speed central processing unit, which has fast calculation speed, high precision and reliable measurement performance. The converter circuit design adopts the latest international technology, the input impedance is up to 1015 ohms, the common mode rejection ratio is better than 100db, and the external interference and 60Hz/50Hz interference suppression ability is better than 90db, which can measure the lower conductivity fluid medium flow. The sensor adopts a new non-uniform magnetic field technology and a special magnetic circuit structure. The magnetic field is stable and reliable, and the volume is greatly reduced, the weight is reduced, and the flowmeter has the characteristics of small size and light weight. Adhering to the enterprise spirit of "seeking truth, hard work, innovation and development", we insist on "customer first, technology first, high quality, first-class service", we will provide you with first-class products and first-class service.