Selection of electrodes and lining
Standard electrodes are used for media that do not produce non-staining electrodes such as crystals, agglomerates, and scales. Otherwise, a doctor blade electrode is used. In addition to the pure alkaline solution, the dosing solution (crystallization of boron and magnesium double salt), the washing liquid and other electromagnetic flowmeters that use the scraper electrode, the other uses the electromagnetic flowmeter of the standard electrode. Since the process pipes in the workshop are basically made of plastic insulation material, if the electromagnetic flowmeter with two electrodes is selected, the grounding ring must be configured to ensure the measurement accuracy. However, the common grounding ring is easy to corrode, and the corrosion-resistant titanium or 钽 grounding ring is high in cost, so the three-electrode electromagnetic flowmeter is selected to ensure measurement accuracy, reduce cost, and convenient installation and maintenance.
Flowmeter type selection
The types of electromagnetic flowmeters are divided into: integral type and split type. In Qinghai, the coldest temperature in winter is up to -30 °C. Many metered electromagnetic flowmeters are outdoors, and the environmental humidity of magnesium-lithium separation, pressure filtration and washing processes in lithium and boron systems is large. Exquisite, evaporation, conversion and other processes up to 90 °C. Therefore, the humidity of the high temperature, low temperature, high temperature fluid, vibration source and other working conditions select a split flow meter to effectively prevent the impact on the measuring components, to ensure measurement accuracy and the life of the electromagnetic flowmeter. However, it should be noted that the signal cable between the electromagnetic flowmeter sensor and the transmitter should not be too long, and the manufacturer-specific cable should be used, otherwise the measurement accuracy will be affected. The rest of the working conditions are more ideal.
The structure of the electromagnetic flowmeter is mainly composed of a magnetic circuit system, a measuring conduit, an electrode, a casing, a lining, and a converter.
Magnetic circuit system: its role is to produce a uniform DC or AC magnetic field. The DC magnetic circuit is realized by a permanent magnet, which has the advantages of simple structure and less interference by the alternating magnetic field, but it is easy to polarize the electrolyte liquid in the measuring duct, so that the positive electrode is surrounded by negative ions, and the negative electrode is positive ion Surrounding, that is, the polarization phenomenon of the electrode, and causing an increase in internal resistance between the two electrodes, thus seriously affecting the normal operation of the meter. When the diameter of the pipe is large, the permanent magnets are correspondingly large, bulky and uneconomical, so the electromagnetic flowmeter generally adopts an alternating magnetic field and is generated by the excitation of a 50HZ power frequency power source.
Measuring catheter: its function is to let the conductive liquid to be tested pass. In order to make the magnetic flux diverted or short-circuited when the magnetic flux passes through the measuring catheter, the measuring catheter must be made of non-magnetic, low electrical conductivity, low thermal conductivity and mechanical strength. Non-magnetic stainless steel, FRP, high strength can be used. Plastic, aluminum, etc.
Use and promotion of electromagnetic flowmeter
The intelligent electromagnetic flowmeter is a fully intelligent electromagnetic flowmeter developed by our company using advanced technology at home and abroad. Its all-Chinese electromagnetic converter core adopts high-speed central processing unit, which has fast calculation speed, high precision and reliable measurement performance. The converter circuit design adopts the latest international technology, the input impedance is up to 1015 ohms, the common mode rejection ratio is better than 100db, and the external interference and 60Hz/50Hz interference suppression ability is better than 90db, which can measure the lower conductivity fluid medium flow. The sensor adopts a new non-uniform magnetic field technology and a special magnetic circuit structure. The magnetic field is stable and reliable, and the volume is greatly reduced, the weight is reduced, and the flowmeter has the characteristics of small size and light weight. Adhering to the enterprise spirit of "seeking truth, hard work, innovation and development", we insist on "customer first, technology first, high quality, first-class service", we will provide you with first-class products and first-class service.