Electromagnetic flowmeter features
Frequency programmable low frequency rectangular wave excitation improves stability of flow measurement and low power loss.
The new 16-bit ultra-low power microprocessor with FLASH memory has high integration, fast calculation speed and high calculation accuracy.
Full digital processing, strong anti-interference ability and reliable measurement.
Ultra-low EMI switching power supply, suitable for power supply voltage variation range, high efficiency, small temperature rise; good EMC performance.
Chinese and English menu operation, easy to use, easy to operate, easy to learn and understand.
High definition backlight wide temperature LCD display.
It can perform bidirectional flow measurement and bidirectional total accumulation; it has automatic range switching function, which can effectively improve the measurement accuracy of analog current and frequency output, especially suitable for occasions where the diurnal flow range changes greatly and needs to send control signals; flow measurement range Up to 1500:1.
There are three totalizers inside, which record and display the forward cumulative amount, the reverse cumulative amount and the accumulated difference integrated amount, which are convenient for fluid metering and custody transfer.
Provide isolated or non-isolated RS485/RS232C digital communication interface, and support fieldbus communication modes such as MODBUS, PROFIBUS-DP and HART.
Constant current source fluid resistance measurement can accurately measure the internal resistance of the electrode signal in the case of long-line transmission. It can be used not only to determine whether the fluid in the sensor is empty or not, but also to identify abnormal phenomena such as contamination and coverage of the electrode, and to provide cleaning for the user. Fault processing information such as electrodes.
Daily maintenance (2)
If it is determined that the converter is faulty, if there is no problem in checking the external cause, please contact the manufacturer of the electromagnetic flowmeter. The manufacturer will generally solve the problem by replacing the circuit board.
1. Before using the electromagnetic flowmeter, first calibrate the electromagnetic flowmeter with a standard pH solution. Before the calibration, before operation, you must pay attention to clean the electrode of the electromagnetic flowmeter with distilled water, and then clean the electrode again with the test liquid.
2. If you do not use the electromagnetic flowmeter, when you want to remove the electrode of the electromagnetic flowmeter, you should be careful not to let the electrode of the electrode collide with the hard object, otherwise the damage will affect the use of the electrode.
3. After using the electromagnetic flowmeter, you should put the electrode of the electromagnetic flowmeter on the sleeve, and put less saturated solution inside, just make sure that the bulb of the electrode is wet, but remember not to put it in distilled water. soak.
4. It is usually necessary to keep the electrode clean and do not let the output on both sides of it appear a short circuit. Otherwise, the measurement will be inaccurate and affect the use of the electromagnetic flowmeter.
In fact, there are still many ways to maintain the electrodes of the electromagnetic flowmeter. Everyone should pay more attention to the use process. Do not use the small negligence of the electromagnetic flowmeter in the future.
Vortex flowmeter selection
(1) Selection of vortex flow transmitter
In the measurement of saturated steam, VA type piezoelectric vortex flow transmitter is adopted. Because of the wide range of vortex flowmeter, in practical applications, it is generally considered that the flow of measuring saturated steam should not be lower than that of vortex flowmeter. The lower limit, that is to say, the fluid flow rate must not be less than 5 m/s. According to the size of the steam, the vortex flow transmitters with different calibers are selected, and the caliber of the transmitter cannot be selected by the existing process pipe diameter.
(2) Selection of pressure compensation pressure transmitter
Since the saturated steam line is long and the pressure fluctuates greatly, pressure compensation must be adopted. Considering the corresponding relationship of pressure, temperature and density, only pressure compensation can be used in the measurement, because the saturated steam pressure of Mingtong Company is 0.3-0.7MPa. Range, the range of pressure transmitter can be selected 1MPa.
(3) Display instrument selection
Display instrument intelligent flow display instrument with voltage regulation compensation, instantaneous flow display and cumulative flow accumulation function.