Plug-in ultrasonic flowmeter features
Programmable frequency low frequency rectangular wave excitation improves stability of flow measurement and low power loss;
Using 16-bit embedded microprocessor, the operation speed is fast and the precision is high;
Full digital processing, strong anti-interference ability, reliable measurement, high precision, flow measurement range up to 100:1;
Ultra-low EMI switching power supply, suitable for wide range of power supply voltage and good anti-EMC performance;
Full Chinese character menu operation, easy to use, easy to operate, easy to learn and understand;
High definition backlit LCD display;
It has two-way flow measurement and two-way total accumulation function, and the current and frequency have bidirectional output function;
There are three totalizers in the internal to display the forward cumulative amount, the reverse cumulative amount and the difference integrated amount;
With RS485 or RS232 digital communication signal output
It has a conductivity measurement function to determine whether the sensor is empty or not;
The constant current excitation current range is large, and can be used with different companies and different types of electromagnetic flow sensors;
With self-test and self-diagnosis function;
High reliability with SMD devices and surface mount (SMT) technology;
The internal design of the meter has a power-down clock that can record 16 power-down times.
Daily maintenance (2)
If it is determined that the converter is faulty, if there is no problem in checking the external cause, please contact the manufacturer of the electromagnetic flowmeter. The manufacturer will generally solve the problem by replacing the circuit board.
1. Before using the electromagnetic flowmeter, first calibrate the electromagnetic flowmeter with a standard pH solution. Before the calibration, before operation, you must pay attention to clean the electrode of the electromagnetic flowmeter with distilled water, and then clean the electrode again with the test liquid.
2. If you do not use the electromagnetic flowmeter, when you want to remove the electrode of the electromagnetic flowmeter, you should be careful not to let the electrode of the electrode collide with the hard object, otherwise the damage will affect the use of the electrode.
3. After using the electromagnetic flowmeter, you should put the electrode of the electromagnetic flowmeter on the sleeve, and put less saturated solution inside, just make sure that the bulb of the electrode is wet, but remember not to put it in distilled water. soak.
4. It is usually necessary to keep the electrode clean and do not let the output on both sides of it appear a short circuit. Otherwise, the measurement will be inaccurate and affect the use of the electromagnetic flowmeter.
In fact, there are still many ways to maintain the electrodes of the electromagnetic flowmeter. Everyone should pay more attention to the use process. Do not use the small negligence of the electromagnetic flowmeter in the future.
Vortex flowmeter verification detection method
Standard table method
The vortex flowmeter is used as a standard device in series with the flowmeter to be inspected, and can be verified by static method or dynamic method. By comparing the readings of the two flow meters, the error of the vortex flowmeter to be tested is obtained.
Standard table flow meter standard device features:
1. Standard table method is suitable for measuring various fluids (including liquids and gases), and also for liquids of various viscosities.
2. The indication value is checked. The flow meter used as the standard meter is installed in series with the flowmeter to be tested in the same closed pipeline system. Generally, there is no time measurement error.
3. As a standard table, the vortex flowmeter can be the same as or different from the flowmeter to be tested.
4. When the flow meter is verified by the standard meter method, the airflow or liquid flow can be not cut off, so it is suitable for online verification, and is also suitable for the measurement standard for the closed pipeline. 5, the standard table method is easy to achieve automation, sealed and safe, does not pollute the environment.
6. Small size, light weight, simple device structure, convenient operation, easy transportation and installation, and low cost.
7. Standard meter flowmeters have low accuracy and poor stability, and often need to be compared regularly or irregularly to monitor their metering performance. The standard table has a shorter verification period.