Because all kinds of ultrasonic flowmeters can be installed outside the pipe, non-contact flow measurement, the cost of the instrument is basically independent of the size of the pipe to be tested, while other types of flowmeters increase with the increase in caliber, so the cost is increased. The flowmeter is superior to the other functions of the same type of flowmeter. It is considered to be a good large-diameter flow measuring instrument. The Doppler ultrasonic flowmeter can measure the flow of two-phase medium, so it can be used for the measurement of dirty sewage such as sewers and sewage. In power plants, the use of portable ultrasonic flowmeters to measure large pipe diameters such as turbine water inflow and turbine circulating water is much more convenient than in the past. Ultrasonic flow juice can also be used for gas measurement. Pipe diameters range from 2cm to 5m, from a few meters wide open channels, culverts to 500m wide rivers.
In addition, the accuracy of the flow measurement of the ultrasonic measuring instrument is almost independent of the temperature, pressure, viscosity, density and other parameters of the measured fluid, and can be made into non-contact and portable measuring instruments, so it can solve the problem that other types of instruments are difficult to measure. Flow measurement problems for corrosive, non-conductive, radioactive, and flammable and explosive media. In addition, in view of the non-contact measurement characteristics, coupled with reasonable electronic circuits, one instrument can adapt to a variety of pipe diameter measurements and a variety of flow range measurements. The adaptability of ultrasonic flowmeters is also unmatched by other instruments. Ultrasonic flowmeters have some of the above advantages, so it has received more and more attention and has been developed into a series of products and generalization. It has been made into standard, high-temperature, explosion-proof and wet instruments of different channels to adapt to different media. Flow measurement for occasions and different pipeline conditions.
Vortex flowmeter common fault
1. The measurement range of the vortex flowmeter is large, generally 10:1, but the lower limit of measurement is limited by many factors: Re>10000 is the most basic condition for the operation of the vortex flowmeter. In addition, it is also responded by the vortex. The signal, the vortex frequency f is also small, and it also makes signal processing difficult. The upper limit of measurement is the frequency response of the sensor and the frequency limit of the circuit. Therefore, the flow rate range must be calculated and calculated according to the flow rate of the fluid. The environmental conditions on the site are complicated. In addition to the conditions such as ambient temperature, humidity, and atmosphere, electromagnetic interference should also be considered.
2, vibration is also a big enemy of this type of instrument. Therefore, care should be taken to avoid mechanical vibrations, especially the lateral vibration of the pipe (perpendicular to the pipe axis and the vertical vortex generating body axis). This effect cannot be suppressed and eliminated in the design of the flowmeter structure. Since the vortex signal is equally sensitive to the influence of the flow field, it is not suitable for the length of the straight pipe section to ensure the flow conditions necessary for stabilizing the vortex street. Even the capacitive and ultrasonic type with strong anti-vibration performance guarantees that the fluid is a fully developed one-way flow, which is not negligible.
The medium temperature also has a great influence on the performance of the vortex flowmeter. For example, the pressure stress type vortex flowmeter cannot be used for a long time at 300 °C, because its insulation resistance is rapidly reduced from 10-100 MΩ at normal temperature to 1-101 Ω, and the output signal is also small, resulting in deterioration of measurement characteristics. In the measurement system, the sensor and the converter should be installed separately to avoid long-term high temperature affecting the reliability and service life of the instrument. The vortex flowmeter is a relatively new type of flowmeter. It is in the development stage and is not very mature. If it is not properly selected, the performance will not work well. Only after reasonable selection and correct installation, it is necessary to carefully and regularly maintain during the use process, accumulate experience, improve the predictability of system failure and the ability to judge and deal with problems, so as to achieve satisfactory results.
Vortex flowmeter verification detection method
Standard table method
The vortex flowmeter is used as a standard device in series with the flowmeter to be inspected, and can be verified by static method or dynamic method. By comparing the readings of the two flow meters, the error of the vortex flowmeter to be tested is obtained.
Standard table flow meter standard device features:
1. Standard table method is suitable for measuring various fluids (including liquids and gases), and also for liquids of various viscosities.
2. The indication value is checked. The flow meter used as the standard meter is installed in series with the flowmeter to be tested in the same closed pipeline system. Generally, there is no time measurement error.
3. As a standard table, the vortex flowmeter can be the same as or different from the flowmeter to be tested.
4. When the flow meter is verified by the standard meter method, the airflow or liquid flow can be not cut off, so it is suitable for online verification, and is also suitable for the measurement standard for the closed pipeline. 5, the standard table method is easy to achieve automation, sealed and safe, does not pollute the environment.
6. Small size, light weight, simple device structure, convenient operation, easy transportation and installation, and low cost.
7. Standard meter flowmeters have low accuracy and poor stability, and often need to be compared regularly or irregularly to monitor their metering performance. The standard table has a shorter verification period.