Choice of protection level
The protection level of the electromagnetic flowmeter should be selected according to the actual situation. If the sensor is required to be installed below the ground and is often flooded, IP68 should be selected. If the sensor is installed above the ground, IP65 and IP67 should be selected. In any case, the display area of ??the electromagnetic flowmeter cannot be used with water or enter the water, which may cause damage to the electromagnetic flowmeter. As a chemical company, the workshop produces high humidity, but IP65 (IP65 is a water-proof type. The electromagnetic flowmeter sensor housing allows the faucet to spray water from the sensor in any direction of the sensor of the electromagnetic flowmeter. The pressure of the water spray is 30Kpa, the water output It is 12.5L/S and the distance is 3 meters.) The electromagnetic flowmeter of the protection grade fully meets the requirements of on-site measurement.
Choice of connection method
Electromagnetic flowmeters have threaded connections, flanged connections, clamps, etc. In the chemical industry, flange-connected electromagnetic flowmeters are generally used. The selection must be consistent with the process-flange engineering pressure and standards.
Conclusion, with the maturity of the domestic instrument industry, chemical companies such as lithium carbonate, boric acid and potash will also enter the market in line with the domestic market. As an important industrial flow measuring instrument, electromagnetic flowmeter should play its role in DCS control. The right selection is especially important to provide accurate measurement data, reduce labor costs and increase production efficiency.
Vortex flowmeter analysis and solution
Summarizing the main causes of these problems, mainly related to the following aspects:
1. Problems with selection. Some vortex sensors are selected on the caliber selection or after the design selection, due to the change of process conditions, so that the selection is larger, the actual selection should be as small as possible to improve the measurement accuracy. The main reason for this is the same. Questions 1, 3, and 6 are related. For example, a vortex pipeline is designed for use by several equipment. Because some of the equipment is not used, the actual actual flow is reduced. The actual design results in too large an original design, which is equivalent to an increase in measurable flow. The lower limit, when the process pipe has a small flow rate, the indication cannot be guaranteed. When the flow rate is large, it can be used, because it is sometimes too difficult to re-engineer. Changes in process conditions are only temporary. The re-tuning of the parameters can be combined to improve the indication accuracy.
2. Installation problems. The main reason is that the length of the straight pipe in front of the sensor is not enough, which affects the measurement accuracy. The reason for this is mainly related to the problem 1. For example, the straight pipe section in front of the sensor is obviously insufficient. Since the FIC203 is not used for measurement, it is only used for control, so the current accuracy can be used equivalent to the downgrade.
Vortex flowmeter analysis and solution
3. Reasons for parameter setting direction. The instrument is incorrectly indicated due to a parameter error. The parameter error makes the secondary meter full frequency calculation error, and the reason for this is mainly related to questions 1 and 3. The full-scale frequency is similar, indicating that the long-term inaccuracy is indicated. The full-scale frequency of the actual full-scale frequency and large-dry calculation indicates that the range is fluctuating and cannot be read. The inconsistency of the parameters on the data affects the final determination of the parameters, and finally passes. Recalibration combined with mutual comparison to determine the parameters solves this problem.
4. The secondary instrument is faulty. There are many faults in this part, including: when the instrument board is disconnected, the range setting has individual bit display bad, and the K coefficient setting has individual bit display bad, which makes it impossible to determine the range setting and K factor setting. Part of the reason is mainly related to questions 1, 2. The problem is solved by fixing the corresponding fault.
5, Four-way line connection problem. On the surface of some circuits, the line connection is very good. Check carefully. Some connectors are actually loose and the circuit is interrupted. Some connectors are tightly connected, but the fastening screws are fastened to the wire due to the secondary line problem. Interruption, this part of the reason is mainly related to question.