﻿ Electro-magnetic flow meter

# Electro-magnetic flow meter

2020-09-26 11:23:44

## Electro-magnetic flow meter

Instruments that measure fluid flow are collectively referred to as flow meters or flow meters. The flowmeter is one of the important instruments in industrial measurement. With the development of industrial production, the accuracy and range of flow measurement requirements are getting higher and higher, and the flow measurement technology is changing with each passing day. Various types of flow meters have been introduced to suit various applications. More than 100 flow meters have been put into use. From different perspectives, flow meters have different classification methods. There are two commonly used classification methods. One is to classify according to the measurement principle adopted by the flowmeter: the second is to classify according to the structural principle of the flowmeter.
Sort by measurement principle
a. Mechanical principle:
Instruments belonging to such principles have differential pressure type, rotor type using Bernoulli's theorem;
Impulse type using the momentum theorem, movable tube type;
Direct mass equation using Newton's second law;
a target using the principle of fluid momentum;
Turbine using the angular momentum theorem;
Vortex type using vortex principle of fluid oscillation;
Use the total static pressure difference of the pitot tube type as well as volumetric and sputum, trough and so on.
b. Electrical principle:
Electromagnetic,
Differential capacitor type,
Inductive,
Strain resistance type, etc.
c. Acoustic principle:
Ultrasonic.
d. Thermal principles:
Thermal type,
Direct calorimetry,
Indirect calorimetry and so on.

## Electro-magnetic flow meter

Vortex flowmeter selection

(1) Selection of vortex flow transmitter
In the measurement of saturated steam, VA type piezoelectric vortex flow transmitter is adopted. Because of the wide range of vortex flowmeter, in practical applications, it is generally considered that the flow of measuring saturated steam should not be lower than that of vortex flowmeter. The lower limit, that is to say, the fluid flow rate must not be less than 5 m/s. According to the size of the steam, the vortex flow transmitters with different calibers are selected, and the caliber of the transmitter cannot be selected by the existing process pipe diameter.
(2) Selection of pressure compensation pressure transmitter
Since the saturated steam line is long and the pressure fluctuates greatly, pressure compensation must be adopted. Considering the corresponding relationship of pressure, temperature and density, only pressure compensation can be used in the measurement, because the saturated steam pressure of Mingtong Company is 0.3-0.7MPa. Range, the range of pressure transmitter can be selected 1MPa.
(3) Display instrument selection
Display instrument intelligent flow display instrument with voltage regulation compensation, instantaneous flow display and cumulative flow accumulation function.

## Electro-magnetic flow meter

Vortex flowmeter analysis and solution

3. Reasons for parameter setting direction. The instrument is incorrectly indicated due to a parameter error. The parameter error makes the secondary meter full frequency calculation error, and the reason for this is mainly related to questions 1 and 3. The full-scale frequency is similar, indicating that the long-term inaccuracy is indicated. The full-scale frequency of the actual full-scale frequency and large-dry calculation indicates that the range is fluctuating and cannot be read. The inconsistency of the parameters on the data affects the final determination of the parameters, and finally passes. Recalibration combined with mutual comparison to determine the parameters solves this problem.
4. The secondary instrument is faulty. There are many faults in this part, including: when the instrument board is disconnected, the range setting has individual bit display bad, and the K coefficient setting has individual bit display bad, which makes it impossible to determine the range setting and K factor setting. Part of the reason is mainly related to questions 1, 2. The problem is solved by fixing the corresponding fault.
5, Four-way line connection problem. On the surface of some circuits, the line connection is very good. Check carefully. Some connectors are actually loose and the circuit is interrupted. Some connectors are tightly connected, but the fastening screws are fastened to the wire due to the secondary line problem. Interruption, this part of the reason is mainly related to question.