Electromagnetic flowmeter features
Frequency programmable low frequency rectangular wave excitation improves stability of flow measurement and low power loss.
The new 16-bit ultra-low power microprocessor with FLASH memory has high integration, fast calculation speed and high calculation accuracy.
Full digital processing, strong anti-interference ability and reliable measurement.
Ultra-low EMI switching power supply, suitable for power supply voltage variation range, high efficiency, small temperature rise; good EMC performance.
Chinese and English menu operation, easy to use, easy to operate, easy to learn and understand.
High definition backlight wide temperature LCD display.
It can perform bidirectional flow measurement and bidirectional total accumulation; it has automatic range switching function, which can effectively improve the measurement accuracy of analog current and frequency output, especially suitable for occasions where the diurnal flow range changes greatly and needs to send control signals; flow measurement range Up to 1500:1.
There are three totalizers inside, which record and display the forward cumulative amount, the reverse cumulative amount and the accumulated difference integrated amount, which are convenient for fluid metering and custody transfer.
Provide isolated or non-isolated RS485/RS232C digital communication interface, and support fieldbus communication modes such as MODBUS, PROFIBUS-DP and HART.
Constant current source fluid resistance measurement can accurately measure the internal resistance of the electrode signal in the case of long-line transmission. It can be used not only to determine whether the fluid in the sensor is empty or not, but also to identify abnormal phenomena such as contamination and coverage of the electrode, and to provide cleaning for the user. Fault processing information such as electrodes.
Vortex flowmeter analysis and solution
6. The connection problem between the secondary instrument and the subsequent instrument. Due to the problem of the subsequent instrument or the maintenance of the subsequent instrument, the mA output circuit of the secondary instrument is interrupted. For this type of secondary instrument, this part is mainly related to the problem 2. Especially for the subsequent recorders, in the case that the recorder cannot be repaired for a long time, it is necessary to pay attention to shorting the output of the secondary meter.
7. The circuit always has no indication due to the failure of the secondary instrument flat-axle cable. Due to long-term operation, coupled with the influence of dust, the flat-axle cable is faulty, and the problem can be solved by cleaning or replacing the flat-axis cable.
8. For the problem 7, the main problem is that the secondary instrument shows that the fixing screw of the meter head is loose, causing the head to sink, the pointer and the case friction are large, the movement is not working, and the problem is solved by adjusting the meter head and re-fixing.
9. Use environmental issues. In particular, the sensor part installed in the well is affected by the humidity of the environment, which causes the circuit board to be damp. This part is mainly related to questions 2 and 2. Through the corresponding technical improvement measures, the sensor part with large humidity is re-separated from the probe part and the conversion part, and the separate type sensor is used. Therefore, the working environment is good, and the instrument has been running well.
10. Due to the poor adjustment of the site, or due to the actual situation after the adjustment. Due to the on-site vibration and noise balance adjustment and sensitivity adjustment is not good. Or because of the re-allocation of the situation after a period of operation after the adjustment, causing the indication problem, this part of the reason is mainly related to questions 4 and 5. Use an oscilloscope, plus the combined process operation, and re-adjust.
Vortex flowmeter installation method
1. The vortex flowmeter can only be measured in one direction. The installation should pay attention to ensure that the direction of the medium flow is consistent with the direction indicated by the flowmeter arrow.
2. The best installation method of the vortex flowmeter is vertical installation, and the medium passes through the flowmeter from bottom to top. Install the flowmeter on a vertical pipe with the flow direction from bottom to top.
3. When installing horizontally, the flowmeter must be installed in the high pressure zone of the whole system and ensure the corresponding outlet pressure; do not install at the highest point of the pipeline, because the highest point is often gas accumulation, the pipeline is not full, and the outlet cannot be directly emptied.
4. When measuring high temperature fluid, try to use vertical installation; if you have to install horizontally, please install the transmitter part of the flowmeter vertically downwards or horizontally to avoid excessive temperature; pay attention to air flow at installation location Or well ventilated.
5. Straight pipe section requirements: at least 15 times the pipe diameter before the flow meter and 5 times the pipe diameter after the flow meter. If there are elbows, indents, expansions and other sources of interference in front of the flowmeter, the diameter of the flowmeter should be 30–40 times, and the diameter of the flowmeter should be 6 times. The flow meter should be installed upstream of the regulator valve, pressure or temperature sensor.
6. When installing, pay attention to the pipe diameter should be slightly larger than or equal to the inner diameter of the instrument.
7. When using the sealing ring, the inner diameter of the sealing ring should be slightly larger than or equal to the inner diameter of the instrument, and the center of the sealing ring is at the center of the pipe.