Ultrasonic flowmeter uses the electronic principle to measure the flow meter. Although it has its own verification procedure, because it has the same usage management mode as the tap water industry measurement, it puts its performance requirements, measurement accuracy, verification cycle, etc. into the cold water meter. In the meantime, it has become more standardized. Since then, the use of large-caliber ultrasonic flowmeters has normative documents in the water industry.
Ultrasonic flowmeter (ultrasonic water meter) Because the measurement principle is different from electromagnetic water meter, its accuracy is different. How to choose an ultrasonic flowmeter with superior performance and precision:
The user flow range is used to determine the type, diameter, temperature, medium, etc. of the ultrasonic flowmeter, especially considering the common flow rate and the starting flow rate. The purpose is to ensure the water supply quantity, accurately measure and reduce the leakage rate. When using the water meter, the user's water meter Commonly used traffic generally cannot exceed the common flow value specified by the product.
Different flow rate to determine the diameter of the flow meter can meet the measurement range.
Different user environments.
Also consider the working pressure, the structure of the flow certificate, the form of the converter (whether bidirectional flow, current, pulse, communication interface, data storage) and the length of the connecting cable.
The performance and accuracy of the ultrasonic flowmeter are selected.
Users can learn more about the peers who are already using the ultrasonic flowmeter, to see the effect of their use, the accuracy of the measurement.
At present, industrial flow measurement generally has problems of large diameter, large flow, small diameter, small flow, high temperature liquid, and gas measurement. This is because the general flowmeter will bring manufacturing and transportation as the diameter of the measuring pipe increases. The difficulty of the above, the advantages of increased cost, increased energy loss, and inconvenient installation, ultrasonic flowmeters can be avoided. The German FLEXIM ultrasonic flowmeter measures diameters from 6mm to 6500 mm, from a few meters wide open channels, culverts to 500m wide rivers.
Vortex flowmeter analysis and solution
6. The connection problem between the secondary instrument and the subsequent instrument. Due to the problem of the subsequent instrument or the maintenance of the subsequent instrument, the mA output circuit of the secondary instrument is interrupted. For this type of secondary instrument, this part is mainly related to the problem 2. Especially for the subsequent recorders, in the case that the recorder cannot be repaired for a long time, it is necessary to pay attention to shorting the output of the secondary meter.
7. The circuit always has no indication due to the failure of the secondary instrument flat-axle cable. Due to long-term operation, coupled with the influence of dust, the flat-axle cable is faulty, and the problem can be solved by cleaning or replacing the flat-axis cable.
8. For the problem 7, the main problem is that the secondary instrument shows that the fixing screw of the meter head is loose, causing the head to sink, the pointer and the case friction are large, the movement is not working, and the problem is solved by adjusting the meter head and re-fixing.
9. Use environmental issues. In particular, the sensor part installed in the well is affected by the humidity of the environment, which causes the circuit board to be damp. This part is mainly related to questions 2 and 2. Through the corresponding technical improvement measures, the sensor part with large humidity is re-separated from the probe part and the conversion part, and the separate type sensor is used. Therefore, the working environment is good, and the instrument has been running well.
10. Due to the poor adjustment of the site, or due to the actual situation after the adjustment. Due to the on-site vibration and noise balance adjustment and sensitivity adjustment is not good. Or because of the re-allocation of the situation after a period of operation after the adjustment, causing the indication problem, this part of the reason is mainly related to questions 4 and 5. Use an oscilloscope, plus the combined process operation, and re-adjust.
Product advantage and preservation
The vortex flowmeter adopts micro-power high-tech, and can be operated continuously by lithium battery for more than one year, which saves the purchase and installation cost of cables and display instruments, and can display instantaneous flow and accumulated flow on the spot. The temperature-compensated integrated vortex flowmeter also has a temperature sensor that directly measures the temperature of the saturated steam and calculates the pressure to show the mass flow of the saturated steam. The temperature and pressure compensation integrated type has a temperature and pressure sensor. The gas flow measurement can directly measure the temperature and pressure of the gas medium, thereby indicating the volumetric volume flow of the gas.
In order to prevent accidental damage to the instrument, please keep the packaging status of our company when it is shipped to the user.
After the instrument arrives, it should be installed in time to avoid the insulation performance of the flow converter being reduced due to unexpected factors, and the metal parts are corroded. If you need to store for a long time, please observe the following:
1. When storing, try not to unpack.
2. The storage location should have the following conditions:
3. With rainproof and waterproof facilities
4. Not susceptible to mechanical vibration or shock
5. The instrument should be stored in the temperature and humidity ranges listed in the table below. The ideal temperature and humidity is 25 ° C, 65%
6. Ambient temperature -20 ° C ~ +60 ° C
7. Relative humidity 5% to 90%