The choice of electromagnetic flowmeter is mainly the correct choice of the transmitter, and the converter only needs to be matched with it.
Choice of caliber and range
The diameter of the transmitter is usually the same as that of the piping system. If the piping system is to be designed, the diameter can be selected according to the flow range and flow rate. For electromagnetic flowmeters, the flow rate is suitable for 2 - 4m / s. In special cases, such as solid particles in the liquid, in consideration of wear, the optional flow rate is ≤ 3m / s, for easy to manage the fluid. Available flow rate ≥ 2m / s. After the flow rate is determined, the transmitter diameter can be determined according to qv= D2.
The range of the transmitter can be selected according to two principles: one is that the full scale of the meter is greater than the expected maximum flow value; the other is that the normal flow is greater than 50% of the full scale of the meter to ensure a certain measurement accuracy.
Temperature and pressure selection
There are certain restrictions on the fluid pressure and temperature that the electromagnetic flowmeter can measure. When used, the pressure used must be lower than the working pressure specified by the flowmeter. At present, the working pressure specifications of domestically produced electromagnetic flowmeters are:
Less than 50mm caliber, working pressure is 1.6MPa; 900 mm caliber, working pressure is 1 MPa;More than 1000mm caliber, working pressure is 0.6MPa.
If there are special requirements on the pressure resistance of the transmitter, you can negotiate with the manufacturer.
The operating temperature of the electromagnetic flowmeter depends on the lining material used, which is generally 5 - 70 ° C. Such as special treatment, can exceed the above range, such as the wear-resistant corrosion-resistant electromagnetic flowmeter produced by Tianjin Automation Instrument No.3. The transmitter allows the measured medium temperature to be -40 to 130 °C.
Selection of lining material and electrode tree material
The lining material and electrode material of the transmitter must be correctly selected according to the physical and chemical properties of the medium. Otherwise, the instrument will be damaged quickly due to the corrosion of the lining and the electrode, and the corrosive medium will easily cause an accident if it leaks. Therefore, the electrode and the lining material must be carefully selected according to the specific measurement medium in the production process.
Vortex flowmeter common fault
1. The measurement range of the vortex flowmeter is large, generally 10:1, but the lower limit of measurement is limited by many factors: Re>10000 is the most basic condition for the operation of the vortex flowmeter. In addition, it is also responded by the vortex. The signal, the vortex frequency f is also small, and it also makes signal processing difficult. The upper limit of measurement is the frequency response of the sensor and the frequency limit of the circuit. Therefore, the flow rate range must be calculated and calculated according to the flow rate of the fluid. The environmental conditions on the site are complicated. In addition to the conditions such as ambient temperature, humidity, and atmosphere, electromagnetic interference should also be considered.
2, vibration is also a big enemy of this type of instrument. Therefore, care should be taken to avoid mechanical vibrations, especially the lateral vibration of the pipe (perpendicular to the pipe axis and the vertical vortex generating body axis). This effect cannot be suppressed and eliminated in the design of the flowmeter structure. Since the vortex signal is equally sensitive to the influence of the flow field, it is not suitable for the length of the straight pipe section to ensure the flow conditions necessary for stabilizing the vortex street. Even the capacitive and ultrasonic type with strong anti-vibration performance guarantees that the fluid is a fully developed one-way flow, which is not negligible.
The medium temperature also has a great influence on the performance of the vortex flowmeter. For example, the pressure stress type vortex flowmeter cannot be used for a long time at 300 °C, because its insulation resistance is rapidly reduced from 10-100 MΩ at normal temperature to 1-101 Ω, and the output signal is also small, resulting in deterioration of measurement characteristics. In the measurement system, the sensor and the converter should be installed separately to avoid long-term high temperature affecting the reliability and service life of the instrument. The vortex flowmeter is a relatively new type of flowmeter. It is in the development stage and is not very mature. If it is not properly selected, the performance will not work well. Only after reasonable selection and correct installation, it is necessary to carefully and regularly maintain during the use process, accumulate experience, improve the predictability of system failure and the ability to judge and deal with problems, so as to achieve satisfactory results.
Product advantage and preservation
The vortex flowmeter adopts micro-power high-tech, and can be operated continuously by lithium battery for more than one year, which saves the purchase and installation cost of cables and display instruments, and can display instantaneous flow and accumulated flow on the spot. The temperature-compensated integrated vortex flowmeter also has a temperature sensor that directly measures the temperature of the saturated steam and calculates the pressure to show the mass flow of the saturated steam. The temperature and pressure compensation integrated type has a temperature and pressure sensor. The gas flow measurement can directly measure the temperature and pressure of the gas medium, thereby indicating the volumetric volume flow of the gas.
In order to prevent accidental damage to the instrument, please keep the packaging status of our company when it is shipped to the user.
After the instrument arrives, it should be installed in time to avoid the insulation performance of the flow converter being reduced due to unexpected factors, and the metal parts are corroded. If you need to store for a long time, please observe the following:
1. When storing, try not to unpack.
2. The storage location should have the following conditions:
3. With rainproof and waterproof facilities
4. Not susceptible to mechanical vibration or shock
5. The instrument should be stored in the temperature and humidity ranges listed in the table below. The ideal temperature and humidity is 25 ° C, 65%
6. Ambient temperature -20 ° C ~ +60 ° C
7. Relative humidity 5% to 90%