To ensure the measurement accuracy of the electromagnetic flowmeter, proper installation is very important.
The transmitter should be installed in a dry and ventilated place indoors. Avoid installation in places where the ambient temperature is too high, should not be subject to strong vibration, try to avoid equipment with strong magnetic fields, such as large motors, transformers, etc. Avoid installation in corrosive gases. The installation location is easy to overhaul. This is the environmental condition to ensure the normal operation of the transmitter.
In order to ensure that the measuring tube of the transmitter is filled with the measured medium, the stacker is preferably installed vertically, and the flow direction is from bottom to top. Especially for liquid-solid two-phase flow, it must be installed vertically. If the site is only allowed to be installed horizontally, it must be ensured that the two electrodes are at the same level.
The transmitter should be equipped with valves and bypass at both ends.
The millivolt AC potential measured by the electrode of the electromagnetic flow transmitter is based on the liquid potential in the transmitter. In order to stabilize the liquid potential and maintain the potential of the transmitter and fluid to ensure stable measurement, the transmitter shell and the metal tube should have good grounding at both ends, and the converter housing should also be grounded. The grounding resistance should not be greater than 10 and cannot be shared with the grounding wire of other electrical equipment. If the transmitter housing is not guaranteed to be in good contact with the metal pipe, connect them with metal wires. Re-reliable grounding.
In order to avoid interference signals, the signal between the transmitter and the converter must be transmitted with shielded wires. It is not allowed to place the signal cable and power cable in parallel in the same cable. The length of the signal cable should generally not exceed 30 m.
Converter installation site should avoid AC and DC strong magnetic field and vibration, ambient temperature is -20 to 50 ° C, does not contain corrosive gases, relative humidity is not more than 80%.
In order to avoid the influence of the flow rate on the relative measurement, the flow regulating valve should be placed downstream of the transmitter. For small-caliber transmitters, since the distance from the center of the electrode to the inlet end of the flowmeter is equivalent to several times the length of the diameter D, the upstream straight pipe may not be specified. However, for a flowmeter with a large diameter, generally there should be a straight pipe section of 5D or more in the upstream, and the downstream pipe section is generally not required.
Vortex flowmeter working principle
The working principle of the vortex flowmeter is to arrange a vortex generator in the fluid, so that the vortex is alternately generated on both sides of the body, and the vortex column is asymmetrically arranged downstream of the vortex generator to generate a certain frequency, by the formula f= St*v/(1-1.27d/D)*d, (St is the Strauhal number, which is a dimensionless number, related to the vortex generator and Reynolds number; v is the flow velocity; d is the incident head width; D is the nominal diameter) to get the flow rate.
In general, the vortex flowmeter output signal (frequency) is not affected by changes in fluid properties and composition, which means that the meter factor is only related to the shape and size of the vortex generator and the Reynolds number. Its advantages are: simple and firm structure, convenient installation and maintenance; suitable for a variety of fluids, liquid, gas, steam and some mixed phases are applicable; high precision, generally up to ± 1% R; flow range is wide, up to 10 : 1 or 20:1 or more; low head loss; no zero drift; relatively cheap price; disadvantage: not suitable for low Reynolds number Re <20000, limited use of high viscosity, low flow rate, small diameter The requirements for the environment are high, and places with vibration should be eliminated as much as possible, and the upstream side needs to have a long straight pipe section; the meter factor is lower, and the larger the diameter, the lower the diameter. The signal resolution is reduced, so the aperture should not be too large, generally used in DN15~DN300mm.
Product advantage and preservation
The vortex flowmeter adopts micro-power high-tech, and can be operated continuously by lithium battery for more than one year, which saves the purchase and installation cost of cables and display instruments, and can display instantaneous flow and accumulated flow on the spot. The temperature-compensated integrated vortex flowmeter also has a temperature sensor that directly measures the temperature of the saturated steam and calculates the pressure to show the mass flow of the saturated steam. The temperature and pressure compensation integrated type has a temperature and pressure sensor. The gas flow measurement can directly measure the temperature and pressure of the gas medium, thereby indicating the volumetric volume flow of the gas.
In order to prevent accidental damage to the instrument, please keep the packaging status of our company when it is shipped to the user.
After the instrument arrives, it should be installed in time to avoid the insulation performance of the flow converter being reduced due to unexpected factors, and the metal parts are corroded. If you need to store for a long time, please observe the following:
1. When storing, try not to unpack.
2. The storage location should have the following conditions:
3. With rainproof and waterproof facilities
4. Not susceptible to mechanical vibration or shock
5. The instrument should be stored in the temperature and humidity ranges listed in the table below. The ideal temperature and humidity is 25 ° C, 65%
6. Ambient temperature -20 ° C ~ +60 ° C
7. Relative humidity 5% to 90%