Selection of electrodes and lining
Standard electrodes are used for media that do not produce non-staining electrodes such as crystals, agglomerates, and scales. Otherwise, a doctor blade electrode is used. In addition to the pure alkaline solution, the dosing solution (crystallization of boron and magnesium double salt), the washing liquid and other electromagnetic flowmeters that use the scraper electrode, the other uses the electromagnetic flowmeter of the standard electrode. Since the process pipes in the workshop are basically made of plastic insulation material, if the electromagnetic flowmeter with two electrodes is selected, the grounding ring must be configured to ensure the measurement accuracy. However, the common grounding ring is easy to corrode, and the corrosion-resistant titanium or 钽 grounding ring is high in cost, so the three-electrode electromagnetic flowmeter is selected to ensure measurement accuracy, reduce cost, and convenient installation and maintenance.
Flowmeter type selection
The types of electromagnetic flowmeters are divided into: integral type and split type. In Qinghai, the coldest temperature in winter is up to -30 °C. Many metered electromagnetic flowmeters are outdoors, and the environmental humidity of magnesium-lithium separation, pressure filtration and washing processes in lithium and boron systems is large. Exquisite, evaporation, conversion and other processes up to 90 °C. Therefore, the humidity of the high temperature, low temperature, high temperature fluid, vibration source and other working conditions select a split flow meter to effectively prevent the impact on the measuring components, to ensure measurement accuracy and the life of the electromagnetic flowmeter. However, it should be noted that the signal cable between the electromagnetic flowmeter sensor and the transmitter should not be too long, and the manufacturer-specific cable should be used, otherwise the measurement accuracy will be affected. The rest of the working conditions are more ideal.
Main features of vortex flowmeter
Measuring medium: nominal diameter of liquid, gas and steam: DN15-DN300 (non-standard products can be customized according to user requirements) Temperature range: -40°C~350°C Pressure specification: PN1.6Mpa; PN2.5Mpa; PN4.0Mpa, Higher pressure specifications can be customized to a specific range: normal range 1:10 Extended range 1:15 Pressure loss factor: Cd ≤ 2.6 System measurement accuracy: liquid, gas indication ± 1%, steam indication ± 1.5% plug-in flow The measured value is ±2.5% of the supply voltage: sensor +12VDC, +24VDC (optional) transmitter +24VDC.
On-site display type The meter comes with 3.6 lithium battery output signal: sensor pulse frequency signal 0.1~3000Hz low level ≤1V high level ≥6V.
Transmitter two-wire 4~20mADC current signal for vibration acceleration: Piezoelectric ≤0.2g Ambient temperature: -40°C~55°C (non-explosion-proof place) -20°C~55°C (explosion-proof place) Ambient humidity: Relative Humidity 5~85% signal remote transmission distance: ≤500m signal line interface: internal thread M20×1.5 explosion-proof grade: iaIICT2-T5 protection grade: ordinary IP65 submersible IP68 instrument material: converter shell is made of aluminum alloy, the body part is adopted 1Cr18Ni9Ti can also be made of special materials according to user requirements.
Product advantage and preservation
The vortex flowmeter adopts micro-power high-tech, and can be operated continuously by lithium battery for more than one year, which saves the purchase and installation cost of cables and display instruments, and can display instantaneous flow and accumulated flow on the spot. The temperature-compensated integrated vortex flowmeter also has a temperature sensor that directly measures the temperature of the saturated steam and calculates the pressure to show the mass flow of the saturated steam. The temperature and pressure compensation integrated type has a temperature and pressure sensor. The gas flow measurement can directly measure the temperature and pressure of the gas medium, thereby indicating the volumetric volume flow of the gas.
In order to prevent accidental damage to the instrument, please keep the packaging status of our company when it is shipped to the user.
After the instrument arrives, it should be installed in time to avoid the insulation performance of the flow converter being reduced due to unexpected factors, and the metal parts are corroded. If you need to store for a long time, please observe the following:
1. When storing, try not to unpack.
2. The storage location should have the following conditions:
3. With rainproof and waterproof facilities
4. Not susceptible to mechanical vibration or shock
5. The instrument should be stored in the temperature and humidity ranges listed in the table below. The ideal temperature and humidity is 25 ° C, 65%
6. Ambient temperature -20 ° C ~ +60 ° C
7. Relative humidity 5% to 90%