Choice of protection level
The protection level of the electromagnetic flowmeter should be selected according to the actual situation. If the sensor is required to be installed below the ground and is often flooded, IP68 should be selected. If the sensor is installed above the ground, IP65 and IP67 should be selected. In any case, the display area of ??the electromagnetic flowmeter cannot be used with water or enter the water, which may cause damage to the electromagnetic flowmeter. As a chemical company, the workshop produces high humidity, but IP65 (IP65 is a water-proof type. The electromagnetic flowmeter sensor housing allows the faucet to spray water from the sensor in any direction of the sensor of the electromagnetic flowmeter. The pressure of the water spray is 30Kpa, the water output It is 12.5L/S and the distance is 3 meters.) The electromagnetic flowmeter of the protection grade fully meets the requirements of on-site measurement.
Choice of connection method
Electromagnetic flowmeters have threaded connections, flanged connections, clamps, etc. In the chemical industry, flange-connected electromagnetic flowmeters are generally used. The selection must be consistent with the process-flange engineering pressure and standards.
Conclusion, with the maturity of the domestic instrument industry, chemical companies such as lithium carbonate, boric acid and potash will also enter the market in line with the domestic market. As an important industrial flow measuring instrument, electromagnetic flowmeter should play its role in DCS control. The right selection is especially important to provide accurate measurement data, reduce labor costs and increase production efficiency.
Vortex flowmeter selection
(1) Selection of vortex flow transmitter
In the measurement of saturated steam, VA type piezoelectric vortex flow transmitter is adopted. Because of the wide range of vortex flowmeter, in practical applications, it is generally considered that the flow of measuring saturated steam should not be lower than that of vortex flowmeter. The lower limit, that is to say, the fluid flow rate must not be less than 5 m/s. According to the size of the steam, the vortex flow transmitters with different calibers are selected, and the caliber of the transmitter cannot be selected by the existing process pipe diameter.
(2) Selection of pressure compensation pressure transmitter
Since the saturated steam line is long and the pressure fluctuates greatly, pressure compensation must be adopted. Considering the corresponding relationship of pressure, temperature and density, only pressure compensation can be used in the measurement, because the saturated steam pressure of Mingtong Company is 0.3-0.7MPa. Range, the range of pressure transmitter can be selected 1MPa.
(3) Display instrument selection
Display instrument intelligent flow display instrument with voltage regulation compensation, instantaneous flow display and cumulative flow accumulation function.
Flow meter type
Instruments that measure fluid flow are collectively referred to as flow meters or flow meters. The flowmeter is one of the important instruments in industrial measurement. With the development of industrial production, the accuracy and range of flow measurement requirements are getting higher and higher, and the flow measurement technology is changing with each passing day. Various types of flow meters have been introduced to suit various applications. More than 100 flow meters have been put into use. From different perspectives, flow meters have different classification methods. There are two commonly used classification methods. One is to classify according to the measurement principle adopted by the flowmeter: the second is to classify according to the structural principle of the flowmeter.
Classified by flowmeter structure principle
A volumetric flow meter is equivalent to a standard volume container that measures the flow medium continuously. The larger the traffic, the more times the metric is, and the higher the frequency of the output. The principle of the volumetric flowmeter is relatively simple and suitable for measuring fluids with high viscosity and low Reynolds number. According to the shape of the rotary body, the products currently produced are: an oval gear flow meter suitable for measuring liquid flow, a lumbar flowmeter (Roots flowmeter), a rotary piston and a scraper flowmeter; a servo type suitable for measuring gas flow Volumetric flowmeters, membranes and flowmeters, etc.