2019/9/21 19:42:40

Choice of caliber

The choice of the diameter of the electromagnetic flowmeter is not necessarily the same as the diameter of the pipe, and should be determined by the flow rate and flow rate. However, as the chemical raw materials and intermediate liquid of Salt Lake, the viscosity is large and the flow rate is low (generally 2.0 to 4.0 m/s). The electromagnetic flowmeter is used on such a pipe, and the diameter of the sensor is the same as the diameter of the pipe. If the flow rate is lower than 1.0m/s, an electromagnetic flowmeter of appropriate flow rate can be selected to ensure measurement accuracy and relatively reduce investment.

Selection of flow rate and range

Basically, they are anti-corrosion plastic pipes, and considering the lining of the flow meter, the flow rate is generally controlled at 2m/s. For some materials that are easy to crystallize (such as sodium carbonate solution, compounding liquid: crystallized mainly boron and magnesium double salt), the flow rate is increased to 3.0 to 4.0 m/s by taking measures. The flow rate will increase the flow noise, and the vibration of the pipeline will affect the measurement accuracy. Under the condition that the electromagnetic flowmeter is installed, the shock absorber should be installed before and after. The full scale of the meter is greater than the expected maximum flow value, which is typically 1.2 times the estimated maximum flow. The normal measurement flow is greater than 50% of the full scale of the meter to ensure a certain measurement accuracy.

Vortex flowmeter common fault

1. The measurement range of the vortex flowmeter is large, generally 10:1, but the lower limit of measurement is limited by many factors: Re>10000 is the most basic condition for the operation of the vortex flowmeter. In addition, it is also responded by the vortex. The signal, the vortex frequency f is also small, and it also makes signal processing difficult. The upper limit of measurement is the frequency response of the sensor and the frequency limit of the circuit. Therefore, the flow rate range must be calculated and calculated according to the flow rate of the fluid. The environmental conditions on the site are complicated. In addition to the conditions such as ambient temperature, humidity, and atmosphere, electromagnetic interference should also be considered.

2, vibration is also a big enemy of this type of instrument. Therefore, care should be taken to avoid mechanical vibrations, especially the lateral vibration of the pipe (perpendicular to the pipe axis and the vertical vortex generating body axis). This effect cannot be suppressed and eliminated in the design of the flowmeter structure. Since the vortex signal is equally sensitive to the influence of the flow field, it is not suitable for the length of the straight pipe section to ensure the flow conditions necessary for stabilizing the vortex street. Even the capacitive and ultrasonic type with strong anti-vibration performance guarantees that the fluid is a fully developed one-way flow, which is not negligible.

The medium temperature also has a great influence on the performance of the vortex flowmeter. For example, the pressure stress type vortex flowmeter cannot be used for a long time at 300 °C, because its insulation resistance is rapidly reduced from 10-100 MΩ at normal temperature to 1-101 Ω, and the output signal is also small, resulting in deterioration of measurement characteristics. In the measurement system, the sensor and the converter should be installed separately to avoid long-term high temperature affecting the reliability and service life of the instrument. The vortex flowmeter is a relatively new type of flowmeter. It is in the development stage and is not very mature. If it is not properly selected, the performance will not work well. Only after reasonable selection and correct installation, it is necessary to carefully and regularly maintain during the use process, accumulate experience, improve the predictability of system failure and the ability to judge and deal with problems, so as to achieve satisfactory results.

Variable area flowmeter (equal pressure drop flowmeter)

The float placed in the upper and lower small tapered flow passages is moved by the force of the fluid flowing from the bottom to the top. When this force is balanced with the "display weight" of the float (the weight of the float itself minus the buoyancy of the fluid it receives), the capture is stationary. The height at which the float is stationary can be used as a measure of the flow rate. Since the cross-sectional area of ??the flowmeter varies with the height of the float, and the pressure difference between the upper and lower parts is equal when the float is stationary, the flowmeter is called a variable area flowmeter or an equal pressure dropmeter. A typical instrument of this type of flow meter is a rotor (float) flow meter.

Momentum flowmeter

A flowmeter weighing flowmeter that uses the momentum of a measuring fluid to reflect the flow rate. Since the momentum P of the flowing fluid is proportional to the density of the fluid and the square of the flow velocity v, i.e., p v2 , when the flow cross section is determined, v is proportional to the volumetric flow rate Q, so p Q2 . Set the proportional coefficient to A, then Q=A. Therefore, P is measured to reflect the flow rate Q. In this type of flowmeter, most of the flowmeters are used to convert momentum into pressure, displacement or force, and then the flow rate is measured. Typical meters for such flow meters are target and rotary wing flow meters.