Ultrasonic flowmeter uses the electronic principle to measure the flow meter. Although it has its own verification procedure, because it has the same usage management mode as the tap water industry measurement, it puts its performance requirements, measurement accuracy, verification cycle, etc. into the cold water meter. In the meantime, it has become more standardized. Since then, the use of large-caliber ultrasonic flowmeters has normative documents in the water industry.
Ultrasonic flowmeter (ultrasonic water meter) Because the measurement principle is different from electromagnetic water meter, its accuracy is different. How to choose an ultrasonic flowmeter with superior performance and precision:
The user flow range is used to determine the type, diameter, temperature, medium, etc. of the ultrasonic flowmeter, especially considering the common flow rate and the starting flow rate. The purpose is to ensure the water supply quantity, accurately measure and reduce the leakage rate. When using the water meter, the user's water meter Commonly used traffic generally cannot exceed the common flow value specified by the product.
Different flow rate to determine the diameter of the flow meter can meet the measurement range.
Different user environments.
Also consider the working pressure, the structure of the flow certificate, the form of the converter (whether bidirectional flow, current, pulse, communication interface, data storage) and the length of the connecting cable.
The performance and accuracy of the ultrasonic flowmeter are selected.
Users can learn more about the peers who are already using the ultrasonic flowmeter, to see the effect of their use, the accuracy of the measurement.
At present, industrial flow measurement generally has problems of large diameter, large flow, small diameter, small flow, high temperature liquid, and gas measurement. This is because the general flowmeter will bring manufacturing and transportation as the diameter of the measuring pipe increases. The difficulty of the above, the advantages of increased cost, increased energy loss, and inconvenient installation, ultrasonic flowmeters can be avoided. The German FLEXIM ultrasonic flowmeter measures diameters from 6mm to 6500 mm, from a few meters wide open channels, culverts to 500m wide rivers.
The working principle of the impeller type flowmeter is that the impeller is placed in the fluid to be measured, and is rotated by the impact of the fluid flow, and the flow rate is reflected by the speed of the impeller rotation. Typical impeller flow meters are water meters and turbine flow meters, which may be of mechanical transmission output or electrical pulse output. Generally, the water meter output of the mechanical transmission has low accuracy and the error is about ±2%, but the structure is simple and the cost is low. The domestic production has been mass-produced, standardized, generalized and serialized. The accuracy of the turbine flowmeter for electrical pulse signal output is high, with a typical error of ±0.2% to 0.5%.
Differential pressure flowmeter (variable pressure drop flowmeter)
The differential pressure flowmeter consists of a primary device and a secondary device. The primary device is called a flow measuring element and is installed in the pipe of the fluid to be measured, generating a pressure difference proportional to the flow rate (flow rate) for the secondary device to display the flow rate. The secondary device is called a display instrument. It receives the differential pressure signal generated by the measuring component and converts it to the corresponding flow for display. The primary device of the differential pressure flow meter is often a throttling device or a dynamic pressure measuring device (piteron, constant velocity tube, etc.). The secondary device is equipped with various mechanical, electronic and combined differential pressure gauges with flow display instruments. The differential pressure sensitive components of the differential pressure gauge are mostly elastic components. Since the differential pressure and the flow rate are in a square root relationship, the flow display instrument is equipped with an open square device to linearize the flow scale. Most meters also have a flow accumulator to display cumulative flow for economic accounting. This method of measuring flow using differential pressure has a long history and is relatively mature. Generally, countries all over the world use it in more important occasions, accounting for about 70% of various flow measurement methods. The flow measurement of the main steam, feed water, condensate, etc. of the power plant is based on this meter.
The flowmeter weighing impulse flowmeter that measures the flow using the impulse theorem is used to measure the flow rate of the granular solid medium, and is also used to measure the flow rate of the mud, the crystalline liquid, and the abrasive. Flow measurement ranges from a few kilograms per hour to nearly 10,000 tons. The typical instrument is a horizontal force-separated impulse flowmeter. The measurement principle is that when the measured medium falls freely from a certain height h to a detection plate with a tilt angle, an impulse is generated, and the horizontal force component of the impulse is proportional to the mass flow rate. Measuring this horizontal component can reflect the mass flow.
The electromagnetic flowmeter is made by applying the electric conductor to generate an induced electromotive force in a magnetic field, and the induced electromotive force is proportional to the flow rate, and is measured by measuring the electromotive force to reflect the flow of the pipeline. Its measurement accuracy and sensitivity are high. Industrially used to measure the flow of water, slurry and other media. The maximum diameter can be measured up to 2m, and the pressure loss is extremely small. However, media with low conductivity, such as gas and steam, cannot be used.
The electromagnetic flowmeter has a high cost, and the signal is susceptible to external magnetic field interference, which affects the wide application in industrial tube flow measurement. To this end, the product is constantly improving and updating, to the development of computer.