The choice of electromagnetic flowmeter is mainly the correct choice of the transmitter, and the converter only needs to be matched with it.
Choice of caliber and range
The diameter of the transmitter is usually the same as that of the piping system. If the piping system is to be designed, the diameter can be selected according to the flow range and flow rate. For electromagnetic flowmeters, the flow rate is suitable for 2 - 4m / s. In special cases, such as solid particles in the liquid, in consideration of wear, the optional flow rate is ≤ 3m / s, for easy to manage the fluid. Available flow rate ≥ 2m / s. After the flow rate is determined, the transmitter diameter can be determined according to qv= D2.
The range of the transmitter can be selected according to two principles: one is that the full scale of the meter is greater than the expected maximum flow value; the other is that the normal flow is greater than 50% of the full scale of the meter to ensure a certain measurement accuracy.
Temperature and pressure selection
There are certain restrictions on the fluid pressure and temperature that the electromagnetic flowmeter can measure. When used, the pressure used must be lower than the working pressure specified by the flowmeter. At present, the working pressure specifications of domestically produced electromagnetic flowmeters are:
Less than 50mm caliber, working pressure is 1.6MPa; 900 mm caliber, working pressure is 1 MPa;More than 1000mm caliber, working pressure is 0.6MPa.
If there are special requirements on the pressure resistance of the transmitter, you can negotiate with the manufacturer.
The operating temperature of the electromagnetic flowmeter depends on the lining material used, which is generally 5 - 70 ° C. Such as special treatment, can exceed the above range, such as the wear-resistant corrosion-resistant electromagnetic flowmeter produced by Tianjin Automation Instrument No.3. The transmitter allows the measured medium temperature to be -40 to 130 °C.
Selection of lining material and electrode tree material
The lining material and electrode material of the transmitter must be correctly selected according to the physical and chemical properties of the medium. Otherwise, the instrument will be damaged quickly due to the corrosion of the lining and the electrode, and the corrosive medium will easily cause an accident if it leaks. Therefore, the electrode and the lining material must be carefully selected according to the specific measurement medium in the production process.
Main features of vortex flowmeter
Measuring medium: nominal diameter of liquid, gas and steam: DN15-DN300 (non-standard products can be customized according to user requirements) Temperature range: -40°C~350°C Pressure specification: PN1.6Mpa; PN2.5Mpa; PN4.0Mpa, Higher pressure specifications can be customized to a specific range: normal range 1:10 Extended range 1:15 Pressure loss factor: Cd ≤ 2.6 System measurement accuracy: liquid, gas indication ± 1%, steam indication ± 1.5% plug-in flow The measured value is ±2.5% of the supply voltage: sensor +12VDC, +24VDC (optional) transmitter +24VDC.
On-site display type The meter comes with 3.6 lithium battery output signal: sensor pulse frequency signal 0.1~3000Hz low level ≤1V high level ≥6V.
Transmitter two-wire 4~20mADC current signal for vibration acceleration: Piezoelectric ≤0.2g Ambient temperature: -40°C~55°C (non-explosion-proof place) -20°C~55°C (explosion-proof place) Ambient humidity: Relative Humidity 5~85% signal remote transmission distance: ≤500m signal line interface: internal thread M20×1.5 explosion-proof grade: iaIICT2-T5 protection grade: ordinary IP65 submersible IP68 instrument material: converter shell is made of aluminum alloy, the body part is adopted 1Cr18Ni9Ti can also be made of special materials according to user requirements.
The flowmeter weighing impulse flowmeter that measures the flow using the impulse theorem is used to measure the flow rate of the granular solid medium, and is also used to measure the flow rate of the mud, the crystalline liquid, and the abrasive. Flow measurement ranges from a few kilograms per hour to nearly 10,000 tons. The typical instrument is a horizontal force-separated impulse flowmeter. The measurement principle is that when the measured medium falls freely from a certain height h to a detection plate with a tilt angle, an impulse is generated, and the horizontal force component of the impulse is proportional to the mass flow rate. Measuring this horizontal component can reflect the mass flow.
The electromagnetic flowmeter is made by applying the electric conductor to generate an induced electromotive force in a magnetic field, and the induced electromotive force is proportional to the flow rate, and is measured by measuring the electromotive force to reflect the flow of the pipeline. Its measurement accuracy and sensitivity are high. Industrially used to measure the flow of water, slurry and other media. The maximum diameter can be measured up to 2m, and the pressure loss is extremely small. However, media with low conductivity, such as gas and steam, cannot be used.
The electromagnetic flowmeter has a high cost, and the signal is susceptible to external magnetic field interference, which affects the wide application in industrial tube flow measurement. To this end, the product is constantly improving and updating, to the development of computer.