Accuracy class and function According to the measurement requirements and the use occasions, the instrument accuracy level is selected to achieve economical efficiency. For example, in the case of trade settlement, product handover and energy measurement, the accuracy level should be higher, such as 1.0, 0.5, or higher; for process control, select different accuracy levels according to control requirements; It is to detect the process flow, no need to do precise control and measurement, you can choose a lower accuracy level, such as 1.5, 2.5, or even 4.0, then you can use a low-cost plug-in electromagnetic flowmeter.
Measuring medium flow rate, meter range and diameter When measuring general medium, the full flow of electromagnetic flowmeter can be selected within the range of 0.5-12m/s of measuring medium flow, and the range is wider. The meter specification (caliber) is not necessarily the same as the process pipeline. It should be determined whether the measured flow range is within the flow rate range. That is, when the pipeline flow rate is too low to meet the flow meter requirements or the measurement accuracy cannot be guaranteed at this flow rate, It is necessary to reduce the gauge diameter, thereby increasing the flow rate inside the tube and obtaining satisfactory measurement results.
Try to avoid ferromagnetic objects and equipment with strong electromagnetic fields to prevent the magnetic field from affecting the working magnetic field and flow signal of the sensor.
Should be installed in the dry and ventilated place, to avoid sun and rain, the ambient temperature should be -20 ~ +60 ° C, relative humidity is less than 85%.
There should be ample space around the flowmeter for easy testing and maintenance.
Vortex flowmeter installation requirements for straight pipe sections:
It is very important that the vortex flowmeter installation meets the requirements for straight pipe sections. Its detailed requirements are as follows:
The flowmeter has certain requirements on the upstream and downstream straight pipe sections at the installation point, otherwise it will affect the measurement accuracy.
If there is a tapered pipe upstream of the installation point of the flowmeter, there should be a straight pipe section of not less than 15D (D is the pipe diameter) upstream of the flowmeter, and a straight pipe section of not less than 5D in the downstream.
If there is a diverging pipe upstream of the installation point of the flowmeter, the upstream of the flowmeter shall have a straight pipe section of not less than 18D (D is the pipe diameter), and the downstream shall have a straight pipe section of not less than 5D.
If there is a 90° elbow or down joint upstream of the installation point of the flowmeter, there should be a straight pipe section of not less than 20D upstream of the flowmeter, and a straight pipe section of not less than 5D downstream.
If there is a 90° elbow on the same plane upstream of the installation point of the flowmeter, there should be a straight pipe section of not less than 25D upstream of the flowmeter, and a straight pipe section of not less than 5D downstream.
The flow regulating valve or pressure regulating valve should be installed as far as possible downstream of the flowmeter 5D. If it must be installed upstream of the flowmeter, the upstream of the flowmeter should have a straight pipe section of not less than 25D, and the downstream should have a straight pipe section of not less than 5D. .
If there is a piston type or plunger type pump in the upstream of the flow meter, a piston type or a Roots type fan and a compressor, the upstream of the flow meter should have a straight pipe section of not less than 25D, and the downstream should have a straight pipe section of not less than 5D.
Special attention: If the valve is installed near the upstream of the installation point of the vortex flowmeter, the valve is continuously opened and closed, which has a great influence on the service life of the flowmeter, and it is very easy to cause permanent damage to the flowmeter. The flowmeter should be avoided to be installed on the very long pipelines in overhead. After a long time, the leakage of the flowmeter can easily cause the leakage of the flowmeter to the flange. If it has to be installed, it must be in the upstream and downstream of the flowmeter. Pipe fastening devices are provided separately.
The V-cone flowmeter is a new generation of differential pressure flow measuring instrument. In practical use, many factors have a great influence on the measurement accuracy of the differential pressure flowmeter, which increases the measurement error and reduces the accuracy.
The specific performance is as follows:
1) the design parameters are inconsistent with the working parameters;
2) there is no temperature compensation or compensation is incorrect;
3) the length of the upper and lower straight pipes is insufficient;
4) the cones and pipes are not concentric;
5) the pressure pipes Blockage, etc.
After on-site inspection, the shortage of straight pipe sections, disagreement, and blockage of the pressure guiding pipe can be ruled out.
The differential pressure transmitter is removed for verification, the transmitter is qualified, and the error is within the allowable range.
The technicians recalled the trend record of DCS preservation, and looked at the steam temperature and pressure recording curve. The temperature was between 120 °C and 150 °C, the pressure was between 0.3MPa and 0.6MPa, and the measured temperature and pressure deviated from the design value. The design parameter was temperature. 193 ° C, pressure 1.13 MPa). At the same time, the DCS configuration was viewed and it was found that there was no temperature and pressure compensation for the steam flow in the configuration.