Use and promotion of electromagnetic flowmeter
The intelligent electromagnetic flowmeter is a fully intelligent electromagnetic flowmeter developed by our company using advanced technology at home and abroad. Its all-Chinese electromagnetic converter core adopts high-speed central processing unit, which has fast calculation speed, high precision and reliable measurement performance. The converter circuit design adopts the latest international technology, the input impedance is up to 1015 ohms, the common mode rejection ratio is better than 100db, and the external interference and 60Hz/50Hz interference suppression ability is better than 90db, which can measure the lower conductivity fluid medium flow. The sensor adopts a new non-uniform magnetic field technology and a special magnetic circuit structure. The magnetic field is stable and reliable, and the volume is greatly reduced, the weight is reduced, and the flowmeter has the characteristics of small size and light weight. Adhering to the enterprise spirit of "seeking truth, hard work, innovation and development", we insist on "customer first, technology first, high quality, first-class service", we will provide you with first-class products and first-class service.
Flow meter type
Instruments that measure fluid flow are collectively referred to as flow meters or flow meters. The flowmeter is one of the important instruments in industrial measurement. With the development of industrial production, the accuracy and range of flow measurement requirements are getting higher and higher, and the flow measurement technology is changing with each passing day. Various types of flow meters have been introduced to suit various applications. More than 100 flow meters have been put into use. From different perspectives, flow meters have different classification methods. There are two commonly used classification methods. One is to classify according to the measurement principle adopted by the flowmeter: the second is to classify according to the structural principle of the flowmeter.
Classified by flowmeter structure principle
A volumetric flow meter is equivalent to a standard volume container that measures the flow medium continuously. The larger the traffic, the more times the metric is, and the higher the frequency of the output. The principle of the volumetric flowmeter is relatively simple and suitable for measuring fluids with high viscosity and low Reynolds number. According to the shape of the rotary body, the products currently produced are: an oval gear flow meter suitable for measuring liquid flow, a lumbar flowmeter (Roots flowmeter), a rotary piston and a scraper flowmeter; a servo type suitable for measuring gas flow Volumetric flowmeters, membranes and flowmeters, etc.
Several problems of steam metering and their solutions
Steam is one of the important energy sources for enterprise production. It is the main heat energy for urban central heating and an important indicator for economic accounting. Therefore, the accuracy of steam metering is particularly important.
In order to solve the problem of heating for employees, the company has updated the original boiler equipment, adding 4 new boilers (2 sets of 25t/h, and another 2 sets of 35t/h). The DCS realizes the automatic control of the boiler system, and the product is saturated steam.
Analysis of problems and influencing factors
Starting from the boiler ignition operation in November 2010, the steam flow often shows the maximum value, and the differential pressure signal measured by the differential pressure transmitter exceeds 20 mA. The same is true when the load is low and the actual flow is small. In this case, steam flow measurement does not provide a safety reference for boiler operation, and it cannot be used for cost accounting.
In response to this phenomenon, the design of the project uses a V-cone flowmeter, equipped with an intelligent differential pressure transmitter, and the measurement signal is sent to the DCS for calculation and display.