Pipe flowmeter measurement method
Ultrasonic waves carry information about the fluid flow rate as they propagate through the flowing fluid. Therefore, the flow rate of the fluid can be detected by the received ultrasonic wave, and converted into a flow rate. According to the detection method, it can be divided into different types of ultrasonic flowmeters such as propagation velocity difference method, Doppler method, beam offset method, noise method and correlation method. Ultrasonic flowmeter is a kind of application that has been applied since the rapid development of integrated circuit technology in the past decade.
Non-contact instrument for measuring fluids that are difficult to access and observe, as well as large pipe runoff. It is linked to a water level gauge for flow measurement of open water flow. The use of ultrasonic flow rate does not change the flow state of the fluid without installing the measuring element in the fluid, and does not generate additional resistance. The installation and maintenance of the instrument can not affect the operation of the production pipeline and is an ideal energy-saving flowmeter.
As we all know, industrial flow measurement generally has the problem of large diameter and large flow measurement difficulty. This is because the general flowmeter will bring difficulties in manufacturing and transportation with the increase of the measuring pipe diameter, and the cost will increase and the energy loss will increase. Installation is not only a disadvantage, but ultrasonic flowmeters can be avoided.
The V-cone flowmeter is a new generation of differential pressure flow measuring instrument. In practical use, many factors have a great influence on the measurement accuracy of the differential pressure flowmeter, which increases the measurement error and reduces the accuracy.
The specific performance is as follows:
1) the design parameters are inconsistent with the working parameters;
2) there is no temperature compensation or compensation is incorrect;
3) the length of the upper and lower straight pipes is insufficient;
4) the cones and pipes are not concentric;
5) the pressure pipes Blockage, etc.
After on-site inspection, the shortage of straight pipe sections, disagreement, and blockage of the pressure guiding pipe can be ruled out.
The differential pressure transmitter is removed for verification, the transmitter is qualified, and the error is within the allowable range.
The technicians recalled the trend record of DCS preservation, and looked at the steam temperature and pressure recording curve. The temperature was between 120 °C and 150 °C, the pressure was between 0.3MPa and 0.6MPa, and the measured temperature and pressure deviated from the design value. The design parameter was temperature. 193 ° C, pressure 1.13 MPa). At the same time, the DCS configuration was viewed and it was found that there was no temperature and pressure compensation for the steam flow in the configuration.
Ultrasonic Flowmeter Introduction and Features
Ultrasonic flow meters are meters that measure the flow by detecting the effect of fluid flow on the ultrasound beam (or ultrasound pulse).
According to the principle of signal detection, ultrasonic flowmeter can be divided into propagation velocity difference method (direct time difference method, time difference method, phase difference method and frequency difference method), beam offset method, Doppler method, cross correlation method, spatial filtering method. And noise law, etc.
Ultrasonic flowmeter is the same as electromagnetic flowmeter. It is an unobstructed flowmeter because it does not have any obstruction parts. It is a kind of flowmeter suitable for solving difficult flow measurement problems, especially in large-diameter flow measurement. The advantage is that it is one of the fastest growing types of flow meters.
Unique signal digital processing technology makes the meter measurement signal more stable, anti-interference ability and more accurate measurement.
No mechanical transmission parts are not easy to damage, maintenance-free and long life.
The circuit is more optimized and integrated; the power consumption is low and the reliability is high.
Intelligent standard signal output, friendly man-machine interface, multiple secondary signal output, for you to choose.
Pipe-type small pipe diameter measurement is economical and convenient, and the measurement accuracy is high.