2019-10-16 08:08:45

Principle analysis of various types of flowmeters

(1) Principles of mechanics: Instruments belonging to such principles have differential pressure and rotor type using Bernoulli's theorem; impulse type and movable tube type using momentum theorem; direct mass type using Newton's second law; The target type of the momentum principle; the turbine type using the angular momentum theorem; the vortex type using the principle of fluid oscillation, the vortex type; the pitot tube type using the total static pressure difference; the volumetric type, the enthalpy, the trough type, and the like.

(2) Electrical principle: The instruments used for such principles are electromagnetic, differential capacitive, inductive, strain resistant, etc.

(3) Acoustic principle: Ultrasonic type, acoustic type (shock wave type), etc. are used for flow measurement using the acoustic principle.

(4) Thermal principle: The heat, direct thermal, indirect calorimetry, etc., which measure the flow using the thermal principle.

(5) Optical principle: laser type, photoelectric type, etc. are instruments belonging to such principles.

(6) Originally based on physical principles: nuclear magnetic resonance, nuclear radiation, etc. are instruments of this type.

(7) Other principles: Marking principle (trace principle, NMR principle), related principles, etc.

Variable area flowmeter (equal pressure drop flowmeter)

The float placed in the upper and lower small tapered flow passages is moved by the force of the fluid flowing from the bottom to the top. When this force is balanced with the "display weight" of the float (the weight of the float itself minus the buoyancy of the fluid it receives), the capture is stationary. The height at which the float is stationary can be used as a measure of the flow rate. Since the cross-sectional area of ??the flowmeter varies with the height of the float, and the pressure difference between the upper and lower parts is equal when the float is stationary, the flowmeter is called a variable area flowmeter or an equal pressure dropmeter. A typical instrument of this type of flow meter is a rotor (float) flow meter.

Momentum flowmeter

A flowmeter weighing flowmeter that uses the momentum of a measuring fluid to reflect the flow rate. Since the momentum P of the flowing fluid is proportional to the density of the fluid and the square of the flow velocity v, i.e., p v2 , when the flow cross section is determined, v is proportional to the volumetric flow rate Q, so p Q2 . Set the proportional coefficient to A, then Q=A. Therefore, P is measured to reflect the flow rate Q. In this type of flowmeter, most of the flowmeters are used to convert momentum into pressure, displacement or force, and then the flow rate is measured. Typical meters for such flow meters are target and rotary wing flow meters.

Ultrasonic Flowmeter Introduction and Features

Definition:

Ultrasonic flow meters are meters that measure the flow by detecting the effect of fluid flow on the ultrasound beam (or ultrasound pulse).

Principle:

According to the principle of signal detection, ultrasonic flowmeter can be divided into propagation velocity difference method (direct time difference method, time difference method, phase difference method and frequency difference method), beam offset method, Doppler method, cross correlation method, spatial filtering method. And noise law, etc.

Ultrasonic flowmeter is the same as electromagnetic flowmeter. It is an unobstructed flowmeter because it does not have any obstruction parts. It is a kind of flowmeter suitable for solving difficult flow measurement problems, especially in large-diameter flow measurement. The advantage is that it is one of the fastest growing types of flow meters.

Features:

Unique signal digital processing technology makes the meter measurement signal more stable, anti-interference ability and more accurate measurement.

No mechanical transmission parts are not easy to damage, maintenance-free and long life.

The circuit is more optimized and integrated; the power consumption is low and the reliability is high.

Intelligent standard signal output, friendly man-machine interface, multiple secondary signal output, for you to choose.

Pipe-type small pipe diameter measurement is economical and convenient, and the measurement accuracy is high.