Electromagnetic flowmeter features
Frequency programmable low frequency rectangular wave excitation improves stability of flow measurement and low power loss.
The new 16-bit ultra-low power microprocessor with FLASH memory has high integration, fast calculation speed and high calculation accuracy.
Full digital processing, strong anti-interference ability and reliable measurement.
Ultra-low EMI switching power supply, suitable for power supply voltage variation range, high efficiency, small temperature rise; good EMC performance.
Chinese and English menu operation, easy to use, easy to operate, easy to learn and understand.
High definition backlight wide temperature LCD display.
It can perform bidirectional flow measurement and bidirectional total accumulation; it has automatic range switching function, which can effectively improve the measurement accuracy of analog current and frequency output, especially suitable for occasions where the diurnal flow range changes greatly and needs to send control signals; flow measurement range Up to 1500:1.
There are three totalizers inside, which record and display the forward cumulative amount, the reverse cumulative amount and the accumulated difference integrated amount, which are convenient for fluid metering and custody transfer.
Provide isolated or non-isolated RS485/RS232C digital communication interface, and support fieldbus communication modes such as MODBUS, PROFIBUS-DP and HART.
Constant current source fluid resistance measurement can accurately measure the internal resistance of the electrode signal in the case of long-line transmission. It can be used not only to determine whether the fluid in the sensor is empty or not, but also to identify abnormal phenomena such as contamination and coverage of the electrode, and to provide cleaning for the user. Fault processing information such as electrodes.
Ultrasonic Flowmeter Introduction and Features
Ultrasonic flow meters are meters that measure the flow by detecting the effect of fluid flow on the ultrasound beam (or ultrasound pulse).
According to the principle of signal detection, ultrasonic flowmeter can be divided into propagation velocity difference method (direct time difference method, time difference method, phase difference method and frequency difference method), beam offset method, Doppler method, cross correlation method, spatial filtering method. And noise law, etc.
Ultrasonic flowmeter is the same as electromagnetic flowmeter. It is an unobstructed flowmeter because it does not have any obstruction parts. It is a kind of flowmeter suitable for solving difficult flow measurement problems, especially in large-diameter flow measurement. The advantage is that it is one of the fastest growing types of flow meters.
Unique signal digital processing technology makes the meter measurement signal more stable, anti-interference ability and more accurate measurement.
No mechanical transmission parts are not easy to damage, maintenance-free and long life.
The circuit is more optimized and integrated; the power consumption is low and the reliability is high.
Intelligent standard signal output, friendly man-machine interface, multiple secondary signal output, for you to choose.
Pipe-type small pipe diameter measurement is economical and convenient, and the measurement accuracy is high.
The correct selection of ultrasonic flowmeters can ensure better use of ultrasonic flowmeters. The type of ultrasonic flowmeter to be selected should be determined according to the physical properties and chemical properties of the fluid medium to be tested, so that the diameter, flow range, lining material, electrode material and output current of the ultrasonic flowmeter can be adapted to the properties of the fluid to be measured. And flow measurement requirements.
1, precision function check
Accuracy levels and functions are based on measurement requirements and usage scenarios to select instrument accuracy levels for cost-effectiveness. For example, in the case of trade settlement, product handover and energy measurement, the accuracy level should be higher, such as 1.0, 0.5, or higher; for process control, select different accuracy levels according to control requirements; It is to detect the process flow, no need to do precise control and measurement, you can choose a lower accuracy level, such as 1.5, 2.5, or even 4.0, then you can use a low-cost plug-in ultrasonic flowmeter.
2, measurable medium
Measuring medium flow rate, meter range and caliber When measuring a general medium, the full flow rate of the ultrasonic flow meter can be selected within the range of 0.5-12 m/s of the measured medium flow rate, and the range is relatively wide. The selection of the meter specification (caliber) is not necessarily the same as the process piping. It should be determined according to whether the measured flow range is within the flow rate range. That is, when the pipeline flow rate is too low to meet the requirements of the flow meter or the measurement accuracy is not guaranteed at this flow rate. It is necessary to reduce the gauge diameter, thereby increasing the flow rate inside the tube and obtaining satisfactory measurement results.