1. Requirements for straight pipe sections
Inlet/outlet straight pipe section: inlet should be ≥10×DN; outlet should be ≥5×DN
2. Grounding point requirement
In order to make the instrument work reliably, improve the measurement accuracy, free from external parasitic potential / the sensor should have good grounding, the grounding resistance is less than 10. (If the metal pipe is well grounded, no special grounding device is required) 3.3 Pair installation Location requirements as shown
Insert the electromagnetic flowmeter according to the situation of the pipeline in the field. If the flowmeter without the ball valve is installed, it should be installed on the pipeline without overpressure (that is, the flowmeter without the ball valve can be installed without pressure), and the hole is opened in the pipeline. Diameter 50, ready to weld the connecting welded pipe to the opening of the pipe; for occasions requiring constant flow loading or unloading or no media spillage, a ball valve must be installed, that is, a plug-in electromagnetic flowmeter with a ball valve structure is selected; With a diameter of 50, it is ready to weld the connecting welded pipe to the opening of the pipe.
Measuring range : Recommended range: 0.5m/s to 10m/s continuously adjustable. Maximum use range: continuously adjustable from 0.2m/s to 15m/s
Signal output: 1, the switch quantity can be set to: pulse output (up to 1000HZ); high/low flow alarm; empty pipe alarm; flow direction indication;
Fault alarm; 2, current output: 4-20mA output
Configuration method: 1. Field configuration through three manual keys. 2. Field configuration via remote control. 3. Perform on-site configuration through the handheld communicator.
Memory: The memory that the EEPROM does not disappear, no battery saving.
Vortex flowmeter selection
(1) Selection of vortex flow transmitter
In the measurement of saturated steam, VA type piezoelectric vortex flow transmitter is adopted. Because of the wide range of vortex flowmeter, in practical applications, it is generally considered that the flow of measuring saturated steam should not be lower than that of vortex flowmeter. The lower limit, that is to say, the fluid flow rate must not be less than 5 m/s. According to the size of the steam, the vortex flow transmitters with different calibers are selected, and the caliber of the transmitter cannot be selected by the existing process pipe diameter.
(2) Selection of pressure compensation pressure transmitter
Since the saturated steam line is long and the pressure fluctuates greatly, pressure compensation must be adopted. Considering the corresponding relationship of pressure, temperature and density, only pressure compensation can be used in the measurement, because the saturated steam pressure of Mingtong Company is 0.3-0.7MPa. Range, the range of pressure transmitter can be selected 1MPa.
(3) Display instrument selection
Display instrument intelligent flow display instrument with voltage regulation compensation, instantaneous flow display and cumulative flow accumulation function.
The correct selection of ultrasonic flowmeters can ensure better use of ultrasonic flowmeters. The type of ultrasonic flowmeter to be selected should be determined according to the physical properties and chemical properties of the fluid medium to be tested, so that the diameter, flow range, lining material, electrode material and output current of the ultrasonic flowmeter can be adapted to the properties of the fluid to be measured. And flow measurement requirements.
1, precision function check
Accuracy levels and functions are based on measurement requirements and usage scenarios to select instrument accuracy levels for cost-effectiveness. For example, in the case of trade settlement, product handover and energy measurement, the accuracy level should be higher, such as 1.0, 0.5, or higher; for process control, select different accuracy levels according to control requirements; It is to detect the process flow, no need to do precise control and measurement, you can choose a lower accuracy level, such as 1.5, 2.5, or even 4.0, then you can use a low-cost plug-in ultrasonic flowmeter.
2, measurable medium
Measuring medium flow rate, meter range and caliber When measuring a general medium, the full flow rate of the ultrasonic flow meter can be selected within the range of 0.5-12 m/s of the measured medium flow rate, and the range is relatively wide. The selection of the meter specification (caliber) is not necessarily the same as the process piping. It should be determined according to whether the measured flow range is within the flow rate range. That is, when the pipeline flow rate is too low to meet the requirements of the flow meter or the measurement accuracy is not guaranteed at this flow rate. It is necessary to reduce the gauge diameter, thereby increasing the flow rate inside the tube and obtaining satisfactory measurement results.