﻿ electro magnetic flowmeter

# electro magnetic flowmeter

2019-10-22 09:42:45

## electro magnetic flowmeter

The electromagnetic flowmeter has a large measuring range, usually 20:1~50:1, and the optional flow range is wide; the diameter of the electromagnetic flowmeter is wider than other types of flowmeters, from a few millimeters to 3 meters; the positive and negative two-way flow can be measured. The pulsating flow rate can also be measured as long as the pulsation frequency is much lower than the excitation frequency; the meter output is linear in nature; it is easy to select the material type of the fluid contact member, and can be applied to corrosive fluids and the like. Since the electromagnetic flowmeter measures much more with suspended solids or dirt than other flow meters, the probability of failure of the inner wall adhesion layer is relatively high. If the electrical conductivity of the adhesion layer is close to the liquid conductivity, the instrument can output the signal normally, but only change the flow area to form a hidden fault of the measurement error; if it is a high conductivity adhesion layer, the electromotive force between the electrodes will be short-circuited; if it is an insulating adhesion layer The surface of the electrode is insulated to disconnect the measuring circuit. The latter two phenomena will make the meter inoperable.

## electro magnetic flowmeter

Vortex flowmeter working principle
The working principle of the vortex flowmeter is to arrange a vortex generator in the fluid, so that the vortex is alternately generated on both sides of the body, and the vortex column is asymmetrically arranged downstream of the vortex generator to generate a certain frequency, by the formula f= St*v/(1-1.27d/D)*d, (St is the Strauhal number, which is a dimensionless number, related to the vortex generator and Reynolds number; v is the flow velocity; d is the incident head width; D is the nominal diameter) to get the flow rate.
In general, the vortex flowmeter output signal (frequency) is not affected by changes in fluid properties and composition, which means that the meter factor is only related to the shape and size of the vortex generator and the Reynolds number. Its advantages are: simple and firm structure, convenient installation and maintenance; suitable for a variety of fluids, liquid, gas, steam and some mixed phases are applicable; high precision, generally up to ± 1% R; flow range is wide, up to 10 : 1 or 20:1 or more; low head loss; no zero drift; relatively cheap price; disadvantage: not suitable for low Reynolds number Re <20000, limited use of high viscosity, low flow rate, small diameter The requirements for the environment are high, and places with vibration should be eliminated as much as possible, and the upstream side needs to have a long straight pipe section; the meter factor is lower, and the larger the diameter, the lower the diameter. The signal resolution is reduced, so the aperture should not be too large, generally used in DN15~DN300mm.

## electro magnetic flowmeter

Ultrasonic flowmeter measurement principle
When the ultrasonic beam propagates in the liquid, the flow of the liquid will cause a small change in the propagation time, and the change in the propagation time is proportional to the flow velocity of the liquid, and its relationship conforms to the following expression.
among them
θ is the angle between the sound beam and the direction of flow of the liquid
M is the number of linear travels of the sound beam in the liquid
D is the inner diameter of the pipe
Tup is the propagation time of the sound beam in the positive direction
Tdown is the propagation time of the sound beam in the reverse direction
ΔT=Tup –Tdown
Let the speed of sound in the stationary fluid be c, the velocity of the fluid flow be u, and the propagation distance be L. When the sound wave is in the same direction as the fluid flow direction (ie, the downstream direction), the propagation velocity is c+u; otherwise, the propagation velocity is cu. Two sets of ultrasonic generators and receivers (T1, R1) and (T2, R2) are placed at two places separated by L. When T1 is in the forward direction and T2 transmits ultrasonic waves in the reverse direction, the time required for the ultrasonic waves to reach the receivers R1 and R2 respectively is t1 and t2, then
T1=L/(c+u); t2=L/(c-u)
Since the flow velocity of the fluid in the industrial pipeline is much smaller than the sound velocity, that is, c>>u, the time difference between the two is ▽t=t2-t1=2Lu/cc. Thus, the propagation velocity of the acoustic wave in the fluid is known. When it is known, the flow rate u can be obtained by measuring the time difference ▽t, and the flow rate Q can be obtained. The method of measuring the flow using this principle is called the time difference method. In addition, a phase difference method, a frequency difference method, or the like can be used.