Ultrasonic flowmeter classification
Plug-in ultrasonic flowmeter: can be installed and maintained without interruption. Ceramic sensors are used for non-stop production installation using dedicated drilling equipment. Generally for mono measurement, in order to improve measurement accuracy, three channels can be selected.
Pipe-type ultrasonic flowmeter: the pipeline installation needs to be cut, but the subsequent maintenance can be stopped. Mono or 3-channel sensors are available.
External clip-on ultrasonic flowmeter: capable of performing fixed and mobile measurements. It is installed with a special coupling agent (silicone rubber cured at room temperature or high-temperature long-chain polymer grease) and does not damage the pipeline during installation.
Portable Ultrasonic Flowmeter: Portable, built-in rechargeable lithium battery, suitable for mobile measurement, with magnetic sensor.
1, non-contact measurement method, small size, easy to carry
2, suitable for on-site measurement of various sizes of pipe sound guiding media
3, built-in nickel-metal hydride rechargeable battery working time of more than 20 hours
4, user interface is flexible, easy to use
5, intelligent on-site printing function to ensure the integrity of the flow data
6, equipped with an integrated aluminum alloy protective box, can be used in harsh outdoor environments
Hand-held ultrasonic flowmeter: small size, light weight, built-in rechargeable lithium battery, hand-held, with magnetic sensor.
Explosion-proof ultrasonic flowmeter: used for explosive liquid flow measurement, it is explosion-proof and intrinsically safe. That is, the converter is explosion-proof and the sensor is intrinsically safe.
Product advantage and preservation
The vortex flowmeter adopts micro-power high-tech, and can be operated continuously by lithium battery for more than one year, which saves the purchase and installation cost of cables and display instruments, and can display instantaneous flow and accumulated flow on the spot. The temperature-compensated integrated vortex flowmeter also has a temperature sensor that directly measures the temperature of the saturated steam and calculates the pressure to show the mass flow of the saturated steam. The temperature and pressure compensation integrated type has a temperature and pressure sensor. The gas flow measurement can directly measure the temperature and pressure of the gas medium, thereby indicating the volumetric volume flow of the gas.
In order to prevent accidental damage to the instrument, please keep the packaging status of our company when it is shipped to the user.
After the instrument arrives, it should be installed in time to avoid the insulation performance of the flow converter being reduced due to unexpected factors, and the metal parts are corroded. If you need to store for a long time, please observe the following:
1. When storing, try not to unpack.
2. The storage location should have the following conditions:
3. With rainproof and waterproof facilities
4. Not susceptible to mechanical vibration or shock
5. The instrument should be stored in the temperature and humidity ranges listed in the table below. The ideal temperature and humidity is 25 ° C, 65%
6. Ambient temperature -20 ° C ~ +60 ° C
7. Relative humidity 5% to 90%
The correct selection of ultrasonic flowmeters can ensure better use of ultrasonic flowmeters. The type of ultrasonic flowmeter to be selected should be determined according to the physical properties and chemical properties of the fluid medium to be tested, so that the diameter, flow range, lining material, electrode material and output current of the ultrasonic flowmeter can be adapted to the properties of the fluid to be measured. And flow measurement requirements.
1, precision function check
Accuracy levels and functions are based on measurement requirements and usage scenarios to select instrument accuracy levels for cost-effectiveness. For example, in the case of trade settlement, product handover and energy measurement, the accuracy level should be higher, such as 1.0, 0.5, or higher; for process control, select different accuracy levels according to control requirements; It is to detect the process flow, no need to do precise control and measurement, you can choose a lower accuracy level, such as 1.5, 2.5, or even 4.0, then you can use a low-cost plug-in ultrasonic flowmeter.
2, measurable medium
Measuring medium flow rate, meter range and caliber When measuring a general medium, the full flow rate of the ultrasonic flow meter can be selected within the range of 0.5-12 m/s of the measured medium flow rate, and the range is relatively wide. The selection of the meter specification (caliber) is not necessarily the same as the process piping. It should be determined according to whether the measured flow range is within the flow rate range. That is, when the pipeline flow rate is too low to meet the requirements of the flow meter or the measurement accuracy is not guaranteed at this flow rate. It is necessary to reduce the gauge diameter, thereby increasing the flow rate inside the tube and obtaining satisfactory measurement results.