To ensure the measurement accuracy of the electromagnetic flowmeter, proper installation is very important.
The transmitter should be installed in a dry and ventilated place indoors. Avoid installation in places where the ambient temperature is too high, should not be subject to strong vibration, try to avoid equipment with strong magnetic fields, such as large motors, transformers, etc. Avoid installation in corrosive gases. The installation location is easy to overhaul. This is the environmental condition to ensure the normal operation of the transmitter.
In order to ensure that the measuring tube of the transmitter is filled with the measured medium, the stacker is preferably installed vertically, and the flow direction is from bottom to top. Especially for liquid-solid two-phase flow, it must be installed vertically. If the site is only allowed to be installed horizontally, it must be ensured that the two electrodes are at the same level.
The transmitter should be equipped with valves and bypass at both ends.
The millivolt AC potential measured by the electrode of the electromagnetic flow transmitter is based on the liquid potential in the transmitter. In order to stabilize the liquid potential and maintain the potential of the transmitter and fluid to ensure stable measurement, the transmitter shell and the metal tube should have good grounding at both ends, and the converter housing should also be grounded. The grounding resistance should not be greater than 10 and cannot be shared with the grounding wire of other electrical equipment. If the transmitter housing is not guaranteed to be in good contact with the metal pipe, connect them with metal wires. Re-reliable grounding.
In order to avoid interference signals, the signal between the transmitter and the converter must be transmitted with shielded wires. It is not allowed to place the signal cable and power cable in parallel in the same cable. The length of the signal cable should generally not exceed 30 m.
Converter installation site should avoid AC and DC strong magnetic field and vibration, ambient temperature is -20 to 50 ° C, does not contain corrosive gases, relative humidity is not more than 80%.
In order to avoid the influence of the flow rate on the relative measurement, the flow regulating valve should be placed downstream of the transmitter. For small-caliber transmitters, since the distance from the center of the electrode to the inlet end of the flowmeter is equivalent to several times the length of the diameter D, the upstream straight pipe may not be specified. However, for a flowmeter with a large diameter, generally there should be a straight pipe section of 5D or more in the upstream, and the downstream pipe section is generally not required.
The V-cone flowmeter is a new generation of differential pressure flow measuring instrument. In practical use, many factors have a great influence on the measurement accuracy of the differential pressure flowmeter, which increases the measurement error and reduces the accuracy.
The specific performance is as follows:
1) the design parameters are inconsistent with the working parameters;
2) there is no temperature compensation or compensation is incorrect;
3) the length of the upper and lower straight pipes is insufficient;
4) the cones and pipes are not concentric;
5) the pressure pipes Blockage, etc.
After on-site inspection, the shortage of straight pipe sections, disagreement, and blockage of the pressure guiding pipe can be ruled out.
The differential pressure transmitter is removed for verification, the transmitter is qualified, and the error is within the allowable range.
The technicians recalled the trend record of DCS preservation, and looked at the steam temperature and pressure recording curve. The temperature was between 120 °C and 150 °C, the pressure was between 0.3MPa and 0.6MPa, and the measured temperature and pressure deviated from the design value. The design parameter was temperature. 193 ° C, pressure 1.13 MPa). At the same time, the DCS configuration was viewed and it was found that there was no temperature and pressure compensation for the steam flow in the configuration.
ltrasonic flowmeter installation
The time difference ultrasonic flowmeter is a competitive flow measurement method in the world today, and its measurement line accuracy is higher than 1.0%. Due to the diversity of the industrial site, especially the environment around the pipeline, how to install and debug the ultrasonic flowmeter according to the specific environment has become an important topic in the field of ultrasonic flow measurement. This procedure explains the installation details of the ultrasonic flowmeter; It further fully demonstrates the advantages of accuracy, reliability and stability of the ultrasonic flowmeter, greatly reducing future maintenance work and even maintenance-free.
Ultrasonic flowmeters should be aware of the site prior to installation, including:
1. What is the distance from the host at the installation of the sensor;
2. Pipe material, pipe wall thickness and pipe diameter;
3. Year of the pipeline;
4. Type of fluid, whether it contains impurities, bubbles, and whether it is full;
5. Fluid temperature
6. Whether there is interference source at the installation site (such as frequency conversion, strong magnetic field, etc.);
7. The temperature of the four seasons at the host site;
8. Whether the power supply voltage used is stable;
9. Whether remote signals and types are required;
According to the site conditions provided above, the manufacturer can configure the site conditions and, if necessary, special models.