1. Requirements for straight pipe sections
Inlet/outlet straight pipe section: inlet should be ≥10×DN; outlet should be ≥5×DN
2. Grounding point requirement
In order to make the instrument work reliably, improve the measurement accuracy, free from external parasitic potential / the sensor should have good grounding, the grounding resistance is less than 10. (If the metal pipe is well grounded, no special grounding device is required) 3.3 Pair installation Location requirements as shown
Insert the electromagnetic flowmeter according to the situation of the pipeline in the field. If the flowmeter without the ball valve is installed, it should be installed on the pipeline without overpressure (that is, the flowmeter without the ball valve can be installed without pressure), and the hole is opened in the pipeline. Diameter 50, ready to weld the connecting welded pipe to the opening of the pipe; for occasions requiring constant flow loading or unloading or no media spillage, a ball valve must be installed, that is, a plug-in electromagnetic flowmeter with a ball valve structure is selected; With a diameter of 50, it is ready to weld the connecting welded pipe to the opening of the pipe.
Measuring range : Recommended range: 0.5m/s to 10m/s continuously adjustable. Maximum use range: continuously adjustable from 0.2m/s to 15m/s
Signal output: 1, the switch quantity can be set to: pulse output (up to 1000HZ); high/low flow alarm; empty pipe alarm; flow direction indication;
Fault alarm; 2, current output: 4-20mA output
Configuration method: 1. Field configuration through three manual keys. 2. Field configuration via remote control. 3. Perform on-site configuration through the handheld communicator.
Memory: The memory that the EEPROM does not disappear, no battery saving.
The flowmeter weighing impulse flowmeter that measures the flow using the impulse theorem is used to measure the flow rate of the granular solid medium, and is also used to measure the flow rate of the mud, the crystalline liquid, and the abrasive. Flow measurement ranges from a few kilograms per hour to nearly 10,000 tons. The typical instrument is a horizontal force-separated impulse flowmeter. The measurement principle is that when the measured medium falls freely from a certain height h to a detection plate with a tilt angle, an impulse is generated, and the horizontal force component of the impulse is proportional to the mass flow rate. Measuring this horizontal component can reflect the mass flow.
The electromagnetic flowmeter is made by applying the electric conductor to generate an induced electromotive force in a magnetic field, and the induced electromotive force is proportional to the flow rate, and is measured by measuring the electromotive force to reflect the flow of the pipeline. Its measurement accuracy and sensitivity are high. Industrially used to measure the flow of water, slurry and other media. The maximum diameter can be measured up to 2m, and the pressure loss is extremely small. However, media with low conductivity, such as gas and steam, cannot be used.
The electromagnetic flowmeter has a high cost, and the signal is susceptible to external magnetic field interference, which affects the wide application in industrial tube flow measurement. To this end, the product is constantly improving and updating, to the development of computer.
The installation position and installation method of the ultrasonic flowmeter.
Selecting the installation pipe segment has a great influence on the test accuracy. The selected pipe segment should avoid the interference and eddy current, which have great influence on the measurement accuracy. Generally, the pipe segment should meet the following conditions:
1. Avoid installing the machine in the pump, high-power radio, frequency conversion, that is, where there is strong magnetic field and vibration interference;
2. Select the pipe section where the pipe should be uniform and dense, and it is easy to transmit ultrasonic waves;
3, to have a long straight pipe section, the upstream straight pipe section of the installation point must be greater than 10D (Note: D = diameter), downstream is greater than 5D;
4, the installation point upstream distance pump should have a distance of 30D;
5. The fluid should be filled with pipes;
6. There should be enough space around the pipeline to facilitate the operation of the on-site personnel. The underground pipeline needs to be a test well. The test well is as follows:
Ultrasonic flowmeters generally have two types of probe installation methods, namely Z method and V method.
However, when D < 200mm and the site condition is one of the following conditions, it can also be installed by the Z method:
1. When the measured fluid has high turbidity, when the V method is used to measure the signal or the signal is weak;
2. When the inner wall of the pipe is lined;
3. When the service life of the pipeline is too long and the inner wall is fouled seriously;
For those with better pipeline conditions, even if D is slightly larger than 200mm, in order to improve the measurement accuracy, the V method can be used for installation.