﻿ oem electromagnetic flow meter

# oem electromagnetic flow meter

2019-11-14 00:11:55

## oem electromagnetic flow meter

The electromagnetic flowmeter has a large measuring range, usually 20:1~50:1, and the optional flow range is wide; the diameter of the electromagnetic flowmeter is wider than other types of flowmeters, from a few millimeters to 3 meters; the positive and negative two-way flow can be measured. The pulsating flow rate can also be measured as long as the pulsation frequency is much lower than the excitation frequency; the meter output is linear in nature; it is easy to select the material type of the fluid contact member, and can be applied to corrosive fluids and the like. Since the electromagnetic flowmeter measures much more with suspended solids or dirt than other flow meters, the probability of failure of the inner wall adhesion layer is relatively high. If the electrical conductivity of the adhesion layer is close to the liquid conductivity, the instrument can output the signal normally, but only change the flow area to form a hidden fault of the measurement error; if it is a high conductivity adhesion layer, the electromotive force between the electrodes will be short-circuited; if it is an insulating adhesion layer The surface of the electrode is insulated to disconnect the measuring circuit. The latter two phenomena will make the meter inoperable.

## oem electromagnetic flow meter

Impulse flowmeter
The flowmeter weighing impulse flowmeter that measures the flow using the impulse theorem is used to measure the flow rate of the granular solid medium, and is also used to measure the flow rate of the mud, the crystalline liquid, and the abrasive. Flow measurement ranges from a few kilograms per hour to nearly 10,000 tons. The typical instrument is a horizontal force-separated impulse flowmeter. The measurement principle is that when the measured medium falls freely from a certain height h to a detection plate with a tilt angle, an impulse is generated, and the horizontal force component of the impulse is proportional to the mass flow rate. Measuring this horizontal component can reflect the mass flow.
Electromagnetic flowmeter
The electromagnetic flowmeter is made by applying the electric conductor to generate an induced electromotive force in a magnetic field, and the induced electromotive force is proportional to the flow rate, and is measured by measuring the electromotive force to reflect the flow of the pipeline. Its measurement accuracy and sensitivity are high. Industrially used to measure the flow of water, slurry and other media. The maximum diameter can be measured up to 2m, and the pressure loss is extremely small. However, media with low conductivity, such as gas and steam, cannot be used.
The electromagnetic flowmeter has a high cost, and the signal is susceptible to external magnetic field interference, which affects the wide application in industrial tube flow measurement. To this end, the product is constantly improving and updating, to the development of computer.

## oem electromagnetic flow meter

ltrasonic flowmeter installation
The time difference ultrasonic flowmeter is a competitive flow measurement method in the world today, and its measurement line accuracy is higher than 1.0%. Due to the diversity of the industrial site, especially the environment around the pipeline, how to install and debug the ultrasonic flowmeter according to the specific environment has become an important topic in the field of ultrasonic flow measurement. This procedure explains the installation details of the ultrasonic flowmeter; It further fully demonstrates the advantages of accuracy, reliability and stability of the ultrasonic flowmeter, greatly reducing future maintenance work and even maintenance-free.
Installation details：
Ultrasonic flowmeters should be aware of the site prior to installation, including:
1. What is the distance from the host at the installation of the sensor;
2. Pipe material, pipe wall thickness and pipe diameter;
3. Year of the pipeline;
4. Type of fluid, whether it contains impurities, bubbles, and whether it is full;
5. Fluid temperature
6. Whether there is interference source at the installation site (such as frequency conversion, strong magnetic field, etc.);
7. The temperature of the four seasons at the host site;
8. Whether the power supply voltage used is stable;
9. Whether remote signals and types are required;
According to the site conditions provided above, the manufacturer can configure the site conditions and, if necessary, special models.