Use and promotion of electromagnetic flowmeter
The intelligent electromagnetic flowmeter is a fully intelligent electromagnetic flowmeter developed by our company using advanced technology at home and abroad. Its all-Chinese electromagnetic converter core adopts high-speed central processing unit, which has fast calculation speed, high precision and reliable measurement performance. The converter circuit design adopts the latest international technology, the input impedance is up to 1015 ohms, the common mode rejection ratio is better than 100db, and the external interference and 60Hz/50Hz interference suppression ability is better than 90db, which can measure the lower conductivity fluid medium flow. The sensor adopts a new non-uniform magnetic field technology and a special magnetic circuit structure. The magnetic field is stable and reliable, and the volume is greatly reduced, the weight is reduced, and the flowmeter has the characteristics of small size and light weight. Adhering to the enterprise spirit of "seeking truth, hard work, innovation and development", we insist on "customer first, technology first, high quality, first-class service", we will provide you with first-class products and first-class service.
Vortex flowmeter verification detection method
Standard table method
The vortex flowmeter is used as a standard device in series with the flowmeter to be inspected, and can be verified by static method or dynamic method. By comparing the readings of the two flow meters, the error of the vortex flowmeter to be tested is obtained.
Standard table flow meter standard device features:
1. Standard table method is suitable for measuring various fluids (including liquids and gases), and also for liquids of various viscosities.
2. The indication value is checked. The flow meter used as the standard meter is installed in series with the flowmeter to be tested in the same closed pipeline system. Generally, there is no time measurement error.
3. As a standard table, the vortex flowmeter can be the same as or different from the flowmeter to be tested.
4. When the flow meter is verified by the standard meter method, the airflow or liquid flow can be not cut off, so it is suitable for online verification, and is also suitable for the measurement standard for the closed pipeline. 5, the standard table method is easy to achieve automation, sealed and safe, does not pollute the environment.
6. Small size, light weight, simple device structure, convenient operation, easy transportation and installation, and low cost.
7. Standard meter flowmeters have low accuracy and poor stability, and often need to be compared regularly or irregularly to monitor their metering performance. The standard table has a shorter verification period.
Some experiences on using ultrasonic flowmeter
1. The flowmeter has two modes, diagonal and reflective. When the reflection mode is not detected, it can be measured in diagonal mode, so that we have been using the reflection mode.
2. The flowmeter requires a relatively high pipe, and the insulation layer must be scraped off, otherwise it cannot be measured. When we measured the air conditioning water system, we cut the insulation layer with a knife to prevent the sensor. After the measurement, the cut insulation layer was glued. The surface of the pipe should be as bright as possible. If it is too rough, it must be sanded.
3. It is difficult to measure when the fluid in the pipeline is not full, so the measuring position should be as straight as possible, away from elbows, valves and other places.
4. The authenticity of the flowmeter reading depends on the signal strength. When the signal is too low, the result is basically unreliable, generally more than 60% or even more.
5. Since the readings may vary greatly, the approach we take is to have the flowmeter read continuously, such as continuously recording one minute of reading and then taking the average.
6. Measuring hot water lines is more difficult than cold water lines. Because the wall temperature of the hot water pipe is high, the coupling agent is easily formed at a high temperature. In addition to the product's own coupling agent, we have tried toothpaste.
The summary of ultrasonic flowmeter measurement accuracy and accuracy needs to be improved.