The mediums that need to be measured in the site are different. Now, a brief introduction is made on the selection of flow meters for different media.
The selection of large water flow meter: suitable for electromagnetic flowmeter, ultrasonic flowmeter. Other applications such as vortex flowmeters, orifice flowmeters, etc. have faded out of this application, and these two types of flow meters are the most important in the industry. In particular, it should be mentioned that the application of Kaifeng Shengda's plug-in flowmeter and electromagnetic flowmeter are better.
Sewage, pulp and other turbid liquid meter selection: electromagnetic flowmeter can be used. However, when choosing an electromagnetic flowmeter, consider that the liquid does not contain more air or bubbles.
The meter with more bubbles of liquid selection: Doppler ultrasonic flowmeter can be selected, using this type of flowmeter to measure the fluid with bubbles, the effect is very good.
Flow meter selection with low conductivity such as pure water and demineralized water: Turbine flow meter is very suitable for measuring such fluids.
Flow meter selection for strong corrosive media such as acid and lye: 1. Electromagnetic flowmeter with acid and alkali lining. 2. clamp type ultrasonic flowmeter.
Selection of flowmeters for mediums with large concentration and large solid particle content such as mortar and electric powder slurry: electromagnetic flowmeter.
Flow meter selection for oil media such as oil and diesel: turbine flow meter and ultrasonic flow meter.
Gas flow meter selection: vortex flowmeter.
Ultrasonic Flowmeter Introduction and Features
Ultrasonic flow meters are meters that measure the flow by detecting the effect of fluid flow on the ultrasound beam (or ultrasound pulse).
According to the principle of signal detection, ultrasonic flowmeter can be divided into propagation velocity difference method (direct time difference method, time difference method, phase difference method and frequency difference method), beam offset method, Doppler method, cross correlation method, spatial filtering method. And noise law, etc.
Ultrasonic flowmeter is the same as electromagnetic flowmeter. It is an unobstructed flowmeter because it does not have any obstruction parts. It is a kind of flowmeter suitable for solving difficult flow measurement problems, especially in large-diameter flow measurement. The advantage is that it is one of the fastest growing types of flow meters.
Unique signal digital processing technology makes the meter measurement signal more stable, anti-interference ability and more accurate measurement.
No mechanical transmission parts are not easy to damage, maintenance-free and long life.
The circuit is more optimized and integrated; the power consumption is low and the reliability is high.
Intelligent standard signal output, friendly man-machine interface, multiple secondary signal output, for you to choose.
Pipe-type small pipe diameter measurement is economical and convenient, and the measurement accuracy is high.
Ultrasonic flowmeter common problems (2)
5, the external clip type ultrasonic flowmeter signal is low?
This depends on the technical content of the instrument itself. After a large number of test cases on the spot, it proves that the long-term pipeline, serious scaling and large diameter, the Aitulel AFTU-2W series is better than other external-clip ultrasonic flowmeters. The signal is very fast and the signal is very stable. .
Solution: For large diameter and serious scaling, it is recommended to use a good quality external clamp type ultrasonic flowmeter. The pipe at the probe installation should be cleaned, and the air between the probe and the workpiece surface should be removed with a coupling agent or coupling piece to make the ultrasonic wave. It can be effectively introduced into the pipeline to ensure sufficient sound intensity transmission on the detection surface.
6. Occasionally, there are bubbles in the measurement medium. Does the ultrasonic flowmeter with the time difference method have any influence?
Today's Aitulel AFTU-2W series external clip type ultrasonic flowmeter has dual mode. When there is air bubble, it can be automatically transferred to Doppler mode to measure. When the bubble disappears, it will automatically turn into the time difference method.
7. The meter cannot be used under strong interference on the spot?
On-site inverter or high voltage cable field strong electromagnetic interference
Recommendation: Keep away from the inverter or high voltage cable field electromagnetic interference
8. What is the minimum diameter of the externally clamped ultrasonic flowmeter on the market? What is the highest temperature?
Answer: Now only the minimum diameter of the AFTU-2W series in the world can be measured to 6mm, the temperature can be measured at 550 °C, such as the measurement of molten salt and heat transfer oil, which is not possible with other brands of flowmeters. of.
9. How to choose a suitable ultrasonic flowmeter?
Answer: Pipe material, pipe wall thickness and pipe diameter; fluid type, contain impurities, air bubbles and full pipe; fluid temperature, flow meter type, portable or fixed online.