﻿ Water Flow Measurement

Water Flow Measurement

2019-11-21 10:19:52

Water Flow Measurement

Instruments that measure fluid flow are collectively referred to as flow meters or flow meters. The flowmeter is one of the important instruments in industrial measurement. With the development of industrial production, the accuracy and range of flow measurement requirements are getting higher and higher, and the flow measurement technology is changing with each passing day. Various types of flow meters have been introduced to suit various applications. More than 100 flow meters have been put into use. From different perspectives, flow meters have different classification methods. There are two commonly used classification methods. One is to classify according to the measurement principle adopted by the flowmeter: the second is to classify according to the structural principle of the flowmeter.
Sort by measurement principle
a. Mechanical principle:
Instruments belonging to such principles have differential pressure type, rotor type using Bernoulli's theorem;
Impulse type using the momentum theorem, movable tube type;
Direct mass equation using Newton's second law;
a target using the principle of fluid momentum;
Turbine using the angular momentum theorem;
Vortex type using vortex principle of fluid oscillation;
Use the total static pressure difference of the pitot tube type as well as volumetric and sputum, trough and so on.
b. Electrical principle:
Electromagnetic,
Differential capacitor type,
Inductive,
Strain resistance type, etc.
c. Acoustic principle:
Ultrasonic.
d. Thermal principles:
Thermal type,
Direct calorimetry,
Indirect calorimetry and so on.

Water Flow Measurement

Precautions
The correct selection of ultrasonic flowmeters can ensure better use of ultrasonic flowmeters. The type of ultrasonic flowmeter to be selected should be determined according to the physical properties and chemical properties of the fluid medium to be tested, so that the diameter, flow range, lining material, electrode material and output current of the ultrasonic flowmeter can be adapted to the properties of the fluid to be measured. And flow measurement requirements.
1, precision function check
Accuracy levels and functions are based on measurement requirements and usage scenarios to select instrument accuracy levels for cost-effectiveness. For example, in the case of trade settlement, product handover and energy measurement, the accuracy level should be higher, such as 1.0, 0.5, or higher; for process control, select different accuracy levels according to control requirements; It is to detect the process flow, no need to do precise control and measurement, you can choose a lower accuracy level, such as 1.5, 2.5, or even 4.0, then you can use a low-cost plug-in ultrasonic flowmeter.
2, measurable medium
Measuring medium flow rate, meter range and caliber When measuring a general medium, the full flow rate of the ultrasonic flow meter can be selected within the range of 0.5-12 m/s of the measured medium flow rate, and the range is relatively wide. The selection of the meter specification (caliber) is not necessarily the same as the process piping. It should be determined according to whether the measured flow range is within the flow rate range. That is, when the pipeline flow rate is too low to meet the requirements of the flow meter or the measurement accuracy is not guaranteed at this flow rate. It is necessary to reduce the gauge diameter, thereby increasing the flow rate inside the tube and obtaining satisfactory measurement results.

Water Flow Measurement

Ultrasonic flowmeter common problems (2)
5, the external clip type ultrasonic flowmeter signal is low?
This depends on the technical content of the instrument itself. After a large number of test cases on the spot, it proves that the long-term pipeline, serious scaling and large diameter, the Aitulel AFTU-2W series is better than other external-clip ultrasonic flowmeters. The signal is very fast and the signal is very stable. .
Solution: For large diameter and serious scaling, it is recommended to use a good quality external clamp type ultrasonic flowmeter. The pipe at the probe installation should be cleaned, and the air between the probe and the workpiece surface should be removed with a coupling agent or coupling piece to make the ultrasonic wave. It can be effectively introduced into the pipeline to ensure sufficient sound intensity transmission on the detection surface.
6. Occasionally, there are bubbles in the measurement medium. Does the ultrasonic flowmeter with the time difference method have any influence?
Today's Aitulel AFTU-2W series external clip type ultrasonic flowmeter has dual mode. When there is air bubble, it can be automatically transferred to Doppler mode to measure. When the bubble disappears, it will automatically turn into the time difference method.
7. The meter cannot be used under strong interference on the spot?
On-site inverter or high voltage cable field strong electromagnetic interference
Recommendation: Keep away from the inverter or high voltage cable field electromagnetic interference
8. What is the minimum diameter of the externally clamped ultrasonic flowmeter on the market? What is the highest temperature?
Answer: Now only the minimum diameter of the AFTU-2W series in the world can be measured to 6mm, the temperature can be measured at 550 °C, such as the measurement of molten salt and heat transfer oil, which is not possible with other brands of flowmeters. of.
9. How to choose a suitable ultrasonic flowmeter?
Answer: Pipe material, pipe wall thickness and pipe diameter; fluid type, contain impurities, air bubbles and full pipe; fluid temperature, flow meter type, portable or fixed online.