Ultrasonic flowmeter features
The unique signal digitization processing technology makes the meter measurement signal more stable, anti-interference ability and more accurate measurement.
No mechanical transmission parts are not easily damaged, maintenance-free and have a long service life.
The circuit is more optimized, with high integration, low power consumption and high reliability.
Intelligent standard signal output, friendly man-machine interface, multiple secondary signal output, for you to choose.
Pipe-section small pipe diameter measurement is economical and convenient, and the measurement accuracy is high.
Detailed installation of ultrasonic flowmeter
Ultrasonic flowmeters should be aware of the site prior to installation, including:
1. What is the distance from the host at the installation of the sensor;
2. Pipe material, pipe wall thickness and pipe diameter;
3. Years of pipeline;
4. The type of fluid, whether it contains impurities, bubbles and whether it is full;
5. Fluid temperature;
6. Whether there is interference source at the installation site (such as frequency conversion, strong magnetic field, etc.);
7. Four seasons temperature at the host place;
8. Whether the power supply voltage used is stable;
9. Do you need remote signals and types;
According to the site conditions provided above, the manufacturer can configure the site conditions and, if necessary, special models.
Vortex flowmeter working principle
The working principle of the vortex flowmeter is to arrange a vortex generator in the fluid, so that the vortex is alternately generated on both sides of the body, and the vortex column is asymmetrically arranged downstream of the vortex generator to generate a certain frequency, by the formula f= St*v/(1-1.27d/D)*d, (St is the Strauhal number, which is a dimensionless number, related to the vortex generator and Reynolds number; v is the flow velocity; d is the incident head width; D is the nominal diameter) to get the flow rate.
In general, the vortex flowmeter output signal (frequency) is not affected by changes in fluid properties and composition, which means that the meter factor is only related to the shape and size of the vortex generator and the Reynolds number. Its advantages are: simple and firm structure, convenient installation and maintenance; suitable for a variety of fluids, liquid, gas, steam and some mixed phases are applicable; high precision, generally up to ± 1% R; flow range is wide, up to 10 : 1 or 20:1 or more; low head loss; no zero drift; relatively cheap price; disadvantage: not suitable for low Reynolds number Re <20000, limited use of high viscosity, low flow rate, small diameter The requirements for the environment are high, and places with vibration should be eliminated as much as possible, and the upstream side needs to have a long straight pipe section; the meter factor is lower, and the larger the diameter, the lower the diameter. The signal resolution is reduced, so the aperture should not be too large, generally used in DN15~DN300mm.
Ultrasonic flowmeter common problems (2)
5, the external clip type ultrasonic flowmeter signal is low?
This depends on the technical content of the instrument itself. After a large number of test cases on the spot, it proves that the long-term pipeline, serious scaling and large diameter, the Aitulel AFTU-2W series is better than other external-clip ultrasonic flowmeters. The signal is very fast and the signal is very stable. .
Solution: For large diameter and serious scaling, it is recommended to use a good quality external clamp type ultrasonic flowmeter. The pipe at the probe installation should be cleaned, and the air between the probe and the workpiece surface should be removed with a coupling agent or coupling piece to make the ultrasonic wave. It can be effectively introduced into the pipeline to ensure sufficient sound intensity transmission on the detection surface.
6. Occasionally, there are bubbles in the measurement medium. Does the ultrasonic flowmeter with the time difference method have any influence?
Today's Aitulel AFTU-2W series external clip type ultrasonic flowmeter has dual mode. When there is air bubble, it can be automatically transferred to Doppler mode to measure. When the bubble disappears, it will automatically turn into the time difference method.
7. The meter cannot be used under strong interference on the spot?
On-site inverter or high voltage cable field strong electromagnetic interference
Recommendation: Keep away from the inverter or high voltage cable field electromagnetic interference
8. What is the minimum diameter of the externally clamped ultrasonic flowmeter on the market? What is the highest temperature?
Answer: Now only the minimum diameter of the AFTU-2W series in the world can be measured to 6mm, the temperature can be measured at 550 °C, such as the measurement of molten salt and heat transfer oil, which is not possible with other brands of flowmeters. of.
9. How to choose a suitable ultrasonic flowmeter?
Answer: Pipe material, pipe wall thickness and pipe diameter; fluid type, contain impurities, air bubbles and full pipe; fluid temperature, flow meter type, portable or fixed online.