Electromagnetic flowmeter features
Using intelligent judgment, the measurement correction setting is not used, and the air traffic control alarm and electrode detection application are more convenient;
Advanced "rough error handling" technology, which can remove fluids such as slurry to measure sharp disturbances, reduce output runout, maintain high precision measurement and make output more stable;
With a fluid density setting, it can display mass flow;
Constant current excitation current range, 125mA, 250mA optional, can be used with different manufacturers, different types of electromagnetic flow sensors;
Control function with remote reset of the totalizer, with contact signal input for starting and stopping accumulation, suitable for total inspection and batch processing applications;
With self-test and self-diagnosis function;
Advanced non-volatile memory for higher circuit reliability and effective protection of setup and measurement parameters;
The meter can be equipped with an unpowered clock and memory for recording the power down time, power-on time and power-down time;
The meter has an optional hour recording function that can store flow and electrode resistance measurements for more than 30 days.
The new keyboard processing method avoids the keyboard operation affecting the measurement, and can enter and return the operation menu to make the parameter setting more convenient;
The total display uses 10-bit decimal 9999999999 full-value carry, which solves the practice of double-word full value 4294967285 (hexadecimal FFFF) carry-in habit;
With wireless transmission, the mesh network is organized by the wireless HART protocol.
The installation position and installation method of the ultrasonic flowmeter.
Selecting the installation pipe segment has a great influence on the test accuracy. The selected pipe segment should avoid the interference and eddy current, which have great influence on the measurement accuracy. Generally, the pipe segment should meet the following conditions:
1. Avoid installing the machine in the pump, high-power radio, frequency conversion, that is, where there is strong magnetic field and vibration interference;
2. Select the pipe section where the pipe should be uniform and dense, and it is easy to transmit ultrasonic waves;
3, to have a long straight pipe section, the upstream straight pipe section of the installation point must be greater than 10D (Note: D = diameter), downstream is greater than 5D;
4, the installation point upstream distance pump should have a distance of 30D;
5. The fluid should be filled with pipes;
6. There should be enough space around the pipeline to facilitate the operation of the on-site personnel. The underground pipeline needs to be a test well. The test well is as follows:
Ultrasonic flowmeters generally have two types of probe installation methods, namely Z method and V method.
However, when D < 200mm and the site condition is one of the following conditions, it can also be installed by the Z method:
1. When the measured fluid has high turbidity, when the V method is used to measure the signal or the signal is weak;
2. When the inner wall of the pipe is lined;
3. When the service life of the pipeline is too long and the inner wall is fouled seriously;
For those with better pipeline conditions, even if D is slightly larger than 200mm, in order to improve the measurement accuracy, the V method can be used for installation.
Turbine Flowmeter Product Introduction
A flow meter that uses a turbine for measurement. It first converts the flow rate to the speed of the turbine and then converts the speed into an electrical signal proportional to the flow. This flow meter is used to detect instantaneous flow and total integrated flow, and its output signal is frequency, which is easy to digitize. In the figure, the induction coil and the permanent magnet are fixed together on the casing. When the ferromagnetic turbine blade passes the magnet, the magnetic resistance of the magnetic circuit changes to generate an induced signal. The signal is amplified and shaped by an amplifier and sent to a counter or frequency meter to display the total integrated flow. At the same time, the pulse frequency is frequency-voltage converted to indicate the instantaneous flow rate. The speed of the impeller is proportional to the flow rate, and the number of revolutions of the impeller is proportional to the total amount flowing. The output of the turbine flow meter is a frequency modulated signal that not only improves the immunity of the detection circuit, but also simplifies the flow detection system. It has a turndown ratio of 10:1 and an accuracy of ±0.2%. Turbine flowmeters with small inertia and small size have a time constant of 0.01 seconds.
Turbine flowmeter is the main type of velocity flowmeter. When the fluid to be measured flows through the turbine flowmeter sensor, under the action of the fluid, the impeller is forced to rotate, and its rotational speed is proportional to the average flow velocity of the pipeline. At the same time, the blade periodicity The magnetic flux generated by the electromagnet is cut and the magnetic flux of the coil is changed. According to the principle of electromagnetic induction, a pulsating potential signal, that is, an electric pulse signal, is generated in the coil, and the frequency of the electric pulsation signal is proportional to the flow rate of the fluid to be measured.