1. Requirements for straight pipe sections
Inlet/outlet straight pipe section: inlet should be ≥10×DN; outlet should be ≥5×DN
2. Grounding point requirement
In order to make the instrument work reliably, improve the measurement accuracy, free from external parasitic potential / the sensor should have good grounding, the grounding resistance is less than 10. (If the metal pipe is well grounded, no special grounding device is required) 3.3 Pair installation Location requirements as shown
Insert the electromagnetic flowmeter according to the situation of the pipeline in the field. If the flowmeter without the ball valve is installed, it should be installed on the pipeline without overpressure (that is, the flowmeter without the ball valve can be installed without pressure), and the hole is opened in the pipeline. Diameter 50, ready to weld the connecting welded pipe to the opening of the pipe; for occasions requiring constant flow loading or unloading or no media spillage, a ball valve must be installed, that is, a plug-in electromagnetic flowmeter with a ball valve structure is selected; With a diameter of 50, it is ready to weld the connecting welded pipe to the opening of the pipe.
Measuring range : Recommended range: 0.5m/s to 10m/s continuously adjustable. Maximum use range: continuously adjustable from 0.2m/s to 15m/s
Signal output: 1, the switch quantity can be set to: pulse output (up to 1000HZ); high/low flow alarm; empty pipe alarm; flow direction indication;
Fault alarm; 2, current output: 4-20mA output
Configuration method: 1. Field configuration through three manual keys. 2. Field configuration via remote control. 3. Perform on-site configuration through the handheld communicator.
Memory: The memory that the EEPROM does not disappear, no battery saving.
Product advantage and preservation
The vortex flowmeter adopts micro-power high-tech, and can be operated continuously by lithium battery for more than one year, which saves the purchase and installation cost of cables and display instruments, and can display instantaneous flow and accumulated flow on the spot. The temperature-compensated integrated vortex flowmeter also has a temperature sensor that directly measures the temperature of the saturated steam and calculates the pressure to show the mass flow of the saturated steam. The temperature and pressure compensation integrated type has a temperature and pressure sensor. The gas flow measurement can directly measure the temperature and pressure of the gas medium, thereby indicating the volumetric volume flow of the gas.
In order to prevent accidental damage to the instrument, please keep the packaging status of our company when it is shipped to the user.
After the instrument arrives, it should be installed in time to avoid the insulation performance of the flow converter being reduced due to unexpected factors, and the metal parts are corroded. If you need to store for a long time, please observe the following:
1. When storing, try not to unpack.
2. The storage location should have the following conditions:
3. With rainproof and waterproof facilities
4. Not susceptible to mechanical vibration or shock
5. The instrument should be stored in the temperature and humidity ranges listed in the table below. The ideal temperature and humidity is 25 ° C, 65%
6. Ambient temperature -20 ° C ~ +60 ° C
7. Relative humidity 5% to 90%
Turbine Flowmeter Product Introduction
A flow meter that uses a turbine for measurement. It first converts the flow rate to the speed of the turbine and then converts the speed into an electrical signal proportional to the flow. This flow meter is used to detect instantaneous flow and total integrated flow, and its output signal is frequency, which is easy to digitize. In the figure, the induction coil and the permanent magnet are fixed together on the casing. When the ferromagnetic turbine blade passes the magnet, the magnetic resistance of the magnetic circuit changes to generate an induced signal. The signal is amplified and shaped by an amplifier and sent to a counter or frequency meter to display the total integrated flow. At the same time, the pulse frequency is frequency-voltage converted to indicate the instantaneous flow rate. The speed of the impeller is proportional to the flow rate, and the number of revolutions of the impeller is proportional to the total amount flowing. The output of the turbine flow meter is a frequency modulated signal that not only improves the immunity of the detection circuit, but also simplifies the flow detection system. It has a turndown ratio of 10:1 and an accuracy of ±0.2%. Turbine flowmeters with small inertia and small size have a time constant of 0.01 seconds.
Turbine flowmeter is the main type of velocity flowmeter. When the fluid to be measured flows through the turbine flowmeter sensor, under the action of the fluid, the impeller is forced to rotate, and its rotational speed is proportional to the average flow velocity of the pipeline. At the same time, the blade periodicity The magnetic flux generated by the electromagnet is cut and the magnetic flux of the coil is changed. According to the principle of electromagnetic induction, a pulsating potential signal, that is, an electric pulse signal, is generated in the coil, and the frequency of the electric pulsation signal is proportional to the flow rate of the fluid to be measured.