Ultrasonic flowmeter classification
Plug-in ultrasonic flowmeter: can be installed and maintained without interruption. Ceramic sensors are used for non-stop production installation using dedicated drilling equipment. Generally for mono measurement, in order to improve measurement accuracy, three channels can be selected.
Pipe-type ultrasonic flowmeter: the pipeline installation needs to be cut, but the subsequent maintenance can be stopped. Mono or 3-channel sensors are available.
External clip-on ultrasonic flowmeter: capable of performing fixed and mobile measurements. It is installed with a special coupling agent (silicone rubber cured at room temperature or high-temperature long-chain polymer grease) and does not damage the pipeline during installation.
Portable Ultrasonic Flowmeter: Portable, built-in rechargeable lithium battery, suitable for mobile measurement, with magnetic sensor.
1, non-contact measurement method, small size, easy to carry
2, suitable for on-site measurement of various sizes of pipe sound guiding media
3, built-in nickel-metal hydride rechargeable battery working time of more than 20 hours
4, user interface is flexible, easy to use
5, intelligent on-site printing function to ensure the integrity of the flow data
6, equipped with an integrated aluminum alloy protective box, can be used in harsh outdoor environments
Hand-held ultrasonic flowmeter: small size, light weight, built-in rechargeable lithium battery, hand-held, with magnetic sensor.
Explosion-proof ultrasonic flowmeter: used for explosive liquid flow measurement, it is explosion-proof and intrinsically safe. That is, the converter is explosion-proof and the sensor is intrinsically safe.
Vortex flowmeter selection
(1) Selection of vortex flow transmitter
In the measurement of saturated steam, VA type piezoelectric vortex flow transmitter is adopted. Because of the wide range of vortex flowmeter, in practical applications, it is generally considered that the flow of measuring saturated steam should not be lower than that of vortex flowmeter. The lower limit, that is to say, the fluid flow rate must not be less than 5 m/s. According to the size of the steam, the vortex flow transmitters with different calibers are selected, and the caliber of the transmitter cannot be selected by the existing process pipe diameter.
(2) Selection of pressure compensation pressure transmitter
Since the saturated steam line is long and the pressure fluctuates greatly, pressure compensation must be adopted. Considering the corresponding relationship of pressure, temperature and density, only pressure compensation can be used in the measurement, because the saturated steam pressure of Mingtong Company is 0.3-0.7MPa. Range, the range of pressure transmitter can be selected 1MPa.
(3) Display instrument selection
Display instrument intelligent flow display instrument with voltage regulation compensation, instantaneous flow display and cumulative flow accumulation function.
Turbine Flowmeter Product Introduction
A flow meter that uses a turbine for measurement. It first converts the flow rate to the speed of the turbine and then converts the speed into an electrical signal proportional to the flow. This flow meter is used to detect instantaneous flow and total integrated flow, and its output signal is frequency, which is easy to digitize. In the figure, the induction coil and the permanent magnet are fixed together on the casing. When the ferromagnetic turbine blade passes the magnet, the magnetic resistance of the magnetic circuit changes to generate an induced signal. The signal is amplified and shaped by an amplifier and sent to a counter or frequency meter to display the total integrated flow. At the same time, the pulse frequency is frequency-voltage converted to indicate the instantaneous flow rate. The speed of the impeller is proportional to the flow rate, and the number of revolutions of the impeller is proportional to the total amount flowing. The output of the turbine flow meter is a frequency modulated signal that not only improves the immunity of the detection circuit, but also simplifies the flow detection system. It has a turndown ratio of 10:1 and an accuracy of ±0.2%. Turbine flowmeters with small inertia and small size have a time constant of 0.01 seconds.
Turbine flowmeter is the main type of velocity flowmeter. When the fluid to be measured flows through the turbine flowmeter sensor, under the action of the fluid, the impeller is forced to rotate, and its rotational speed is proportional to the average flow velocity of the pipeline. At the same time, the blade periodicity The magnetic flux generated by the electromagnet is cut and the magnetic flux of the coil is changed. According to the principle of electromagnetic induction, a pulsating potential signal, that is, an electric pulse signal, is generated in the coil, and the frequency of the electric pulsation signal is proportional to the flow rate of the fluid to be measured.