Use and promotion of electromagnetic flowmeter
The intelligent electromagnetic flowmeter is a fully intelligent electromagnetic flowmeter developed by our company using advanced technology at home and abroad. Its all-Chinese electromagnetic converter core adopts high-speed central processing unit, which has fast calculation speed, high precision and reliable measurement performance. The converter circuit design adopts the latest international technology, the input impedance is up to 1015 ohms, the common mode rejection ratio is better than 100db, and the external interference and 60Hz/50Hz interference suppression ability is better than 90db, which can measure the lower conductivity fluid medium flow. The sensor adopts a new non-uniform magnetic field technology and a special magnetic circuit structure. The magnetic field is stable and reliable, and the volume is greatly reduced, the weight is reduced, and the flowmeter has the characteristics of small size and light weight. Adhering to the enterprise spirit of "seeking truth, hard work, innovation and development", we insist on "customer first, technology first, high quality, first-class service", we will provide you with first-class products and first-class service.
Daily maintenance (2)
If it is determined that the converter is faulty, if there is no problem in checking the external cause, please contact the manufacturer of the electromagnetic flowmeter. The manufacturer will generally solve the problem by replacing the circuit board.
1. Before using the electromagnetic flowmeter, first calibrate the electromagnetic flowmeter with a standard pH solution. Before the calibration, before operation, you must pay attention to clean the electrode of the electromagnetic flowmeter with distilled water, and then clean the electrode again with the test liquid.
2. If you do not use the electromagnetic flowmeter, when you want to remove the electrode of the electromagnetic flowmeter, you should be careful not to let the electrode of the electrode collide with the hard object, otherwise the damage will affect the use of the electrode.
3. After using the electromagnetic flowmeter, you should put the electrode of the electromagnetic flowmeter on the sleeve, and put less saturated solution inside, just make sure that the bulb of the electrode is wet, but remember not to put it in distilled water. soak.
4. It is usually necessary to keep the electrode clean and do not let the output on both sides of it appear a short circuit. Otherwise, the measurement will be inaccurate and affect the use of the electromagnetic flowmeter.
In fact, there are still many ways to maintain the electrodes of the electromagnetic flowmeter. Everyone should pay more attention to the use process. Do not use the small negligence of the electromagnetic flowmeter in the future.
Turbine Flowmeter Product Introduction
A flow meter that uses a turbine for measurement. It first converts the flow rate to the speed of the turbine and then converts the speed into an electrical signal proportional to the flow. This flow meter is used to detect instantaneous flow and total integrated flow, and its output signal is frequency, which is easy to digitize. In the figure, the induction coil and the permanent magnet are fixed together on the casing. When the ferromagnetic turbine blade passes the magnet, the magnetic resistance of the magnetic circuit changes to generate an induced signal. The signal is amplified and shaped by an amplifier and sent to a counter or frequency meter to display the total integrated flow. At the same time, the pulse frequency is frequency-voltage converted to indicate the instantaneous flow rate. The speed of the impeller is proportional to the flow rate, and the number of revolutions of the impeller is proportional to the total amount flowing. The output of the turbine flow meter is a frequency modulated signal that not only improves the immunity of the detection circuit, but also simplifies the flow detection system. It has a turndown ratio of 10:1 and an accuracy of ±0.2%. Turbine flowmeters with small inertia and small size have a time constant of 0.01 seconds.
Turbine flowmeter is the main type of velocity flowmeter. When the fluid to be measured flows through the turbine flowmeter sensor, under the action of the fluid, the impeller is forced to rotate, and its rotational speed is proportional to the average flow velocity of the pipeline. At the same time, the blade periodicity The magnetic flux generated by the electromagnet is cut and the magnetic flux of the coil is changed. According to the principle of electromagnetic induction, a pulsating potential signal, that is, an electric pulse signal, is generated in the coil, and the frequency of the electric pulsation signal is proportional to the flow rate of the fluid to be measured.