Electrode: Its function is to extract and induce a proportional induced inductive potential signal. The electrodes are typically made of non-magnetically conductive stainless steel and are required to be flush with the liner so that the fluid passes unimpeded. It should be installed in the vertical direction of the pipe to prevent deposits from accumulating on it and affecting the measurement accuracy.
Enclosure: Made of ferromagnetic material, it is the cover of the distribution system excitation coil and isolates the interference of external magnetic field.
Lining: A complete electrical insulation lining on the inside of the measuring tube and on the flange sealing surface. It directly contacts the liquid to be measured, and its function is to increase the corrosion resistance of the measuring catheter and prevent the induced potential from being short-circuited by the metal measuring tube wall. Most of the lining materials are PTFE plastics and ceramics that are resistant to corrosion, high temperature and wear.
Converter: The induced potential signal generated by the liquid flow is very weak and is greatly affected by various interference factors. The function of the converter is to amplify and convert the induced potential signal into a unified standard signal and suppress the main interference signal. Its task is to amplify the induced potential signal Ex detected by the electrode into a unified standard DC signal.
Executive Standard: JB/T 9248-1999
Nominal diameter: 15, 20, 25, 32, 40, 50, 65, 80, 100, 125, 150, 200, 250, 300, 350, 400, 450, 500, 600, 700, 800, 900, 1000, 1200, 1400, 1600, 1800, 2000, 2200, 2400, 2600, 2800, 3000
Maximum flow rate: 15m/s
Accuracy: DNl5~DN600, indication: ±0.3% (flow rate ≥1m/s); ±3mm/s (flow rate <1m/s)
DN700-DN3000, ±0.5% of the indicated value (flow rate ≥0.8m/S); ±4mm/s (flow rate <0.8m/S)
Fluid conductivity ≥5uS/cm
DNl5～DN: 1504.0MPa, DNl5～DN600: 1.6MPa, DN200～DN1000: 1.0MPa, DN700～DN3000: 0.6MPa, Special order: 6.3, 10MPa
Sensor: -25 °C - ten 60 °C
Converter and integrated type: -10 ° C - ten 60 °C
Lining material: PTFE, neoprene, polyurethane, polytetrafluoroethylene (F46), screened PFA
Maximum fluid temperature
- Body type 70 °C
Separate type: Polychloroprene lining 80 ° C; 120 °C (specify when ordering)
Polyurethane lining 80 °C
Polytetrafluoroethylene propylene (F46) 100 °C; 150 °C (specify when ordering)
Signal electrode and grounding electrode material: stainless steel 0Crl8Nil2M02Ti, Hastelloy C, Hastelloy B, titanium, tantalum, platinum/rhodium alloy, stainless steel coated tungsten carbide
Electrode scraper mechanism: DN300-DN3000
Connecting flange material: carbon steel
Grounding flange material: stainless steel 1Crl8Ni9Ti
DN65—DNl50: Stainless steel 1Crl8Ni9Ti
DN200～DNl600: Carbon steel ten stainless steel 1Crl8Ni9Ti
DNl5～DN3000 separate rubber or polyurethane lining sensor: IP65 or IP68
Other sensors, body flow meters and split converters: IP65
Spacing (separate type): The converter distance sensor generally does not exceed 100m
Gas turbine flowmeter and screwing vortex flowmeter measurement difference
Keywords: gas turbine flowmeter, Tianxin gas turbine flowmeter, gas gas turbine flowmeter
First, the characteristics of gas turbine flowmeter and spiral vortex flowmeter
Gas turbine flowmeter and screwing vortex flowmeter are the main gas metering instruments in China. In foreign countries, almost no one uses gas to swirl into the vortex flow meter. At present, there are certain applications in the domestic vortex flowmeter. In terms of standard conditions. The international standard ISO9551 promulgated by the International Organization for Standardization and the international recommendations R6, R32 promulgated by the International Organization of Legal Metrology (OIML) have corresponding national standards and regulations in China and developed countries. The above documents guarantee the authority of gas turbine flowmeters in gas metering. Sex. Gas-introducing vortex flowmeters have not yet established relevant standards. It is difficult to guarantee the authority of the flow meter in gas metering. At present, in domestic flow meters for gas measurement, gas turbine flowmeters dominate, and the vortex flowmeters also have a certain market. The use of the two is similar, and users often have doubts about the choice of the two.