Ultrasonic flowmeter uses the electronic principle to measure the flow meter. Although it has its own verification procedure, because it has the same usage management mode as the tap water industry measurement, it puts its performance requirements, measurement accuracy, verification cycle, etc. into the cold water meter. In the meantime, it has become more standardized. Since then, the use of large-caliber ultrasonic flowmeters has normative documents in the water industry.
Ultrasonic flowmeter (ultrasonic water meter) Because the measurement principle is different from electromagnetic water meter, its accuracy is different. How to choose an ultrasonic flowmeter with superior performance and precision:
The user flow range is used to determine the type, diameter, temperature, medium, etc. of the ultrasonic flowmeter, especially considering the common flow rate and the starting flow rate. The purpose is to ensure the water supply quantity, accurately measure and reduce the leakage rate. When using the water meter, the user's water meter Commonly used traffic generally cannot exceed the common flow value specified by the product.
Different flow rate to determine the diameter of the flow meter can meet the measurement range.
Different user environments.
Also consider the working pressure, the structure of the flow certificate, the form of the converter (whether bidirectional flow, current, pulse, communication interface, data storage) and the length of the connecting cable.
The performance and accuracy of the ultrasonic flowmeter are selected.
Users can learn more about the peers who are already using the ultrasonic flowmeter, to see the effect of their use, the accuracy of the measurement.
At present, industrial flow measurement generally has problems of large diameter, large flow, small diameter, small flow, high temperature liquid, and gas measurement. This is because the general flowmeter will bring manufacturing and transportation as the diameter of the measuring pipe increases. The difficulty of the above, the advantages of increased cost, increased energy loss, and inconvenient installation, ultrasonic flowmeters can be avoided. The German FLEXIM ultrasonic flowmeter measures diameters from 6mm to 6500 mm, from a few meters wide open channels, culverts to 500m wide rivers.
Executive Standard: JB/T 9248-1999
Nominal diameter: 15, 20, 25, 32, 40, 50, 65, 80, 100, 125, 150, 200, 250, 300, 350, 400, 450, 500, 600, 700, 800, 900, 1000, 1200, 1400, 1600, 1800, 2000, 2200, 2400, 2600, 2800, 3000
Maximum flow rate: 15m/s
Accuracy: DNl5~DN600, indication: ±0.3% (flow rate ≥1m/s); ±3mm/s (flow rate <1m/s)
DN700-DN3000, ±0.5% of the indicated value (flow rate ≥0.8m/S); ±4mm/s (flow rate <0.8m/S)
Fluid conductivity ≥5uS/cm
DNl5～DN: 1504.0MPa, DNl5～DN600: 1.6MPa, DN200～DN1000: 1.0MPa, DN700～DN3000: 0.6MPa, Special order: 6.3, 10MPa
Sensor: -25 °C - ten 60 °C
Converter and integrated type: -10 ° C - ten 60 °C
Lining material: PTFE, neoprene, polyurethane, polytetrafluoroethylene (F46), screened PFA
Maximum fluid temperature
- Body type 70 °C
Separate type: Polychloroprene lining 80 ° C; 120 °C (specify when ordering)
Polyurethane lining 80 °C
Polytetrafluoroethylene propylene (F46) 100 °C; 150 °C (specify when ordering)
Signal electrode and grounding electrode material: stainless steel 0Crl8Nil2M02Ti, Hastelloy C, Hastelloy B, titanium, tantalum, platinum/rhodium alloy, stainless steel coated tungsten carbide
Electrode scraper mechanism: DN300-DN3000
Connecting flange material: carbon steel
Grounding flange material: stainless steel 1Crl8Ni9Ti
DN65—DNl50: Stainless steel 1Crl8Ni9Ti
DN200～DNl600: Carbon steel ten stainless steel 1Crl8Ni9Ti
DNl5～DN3000 separate rubber or polyurethane lining sensor: IP65 or IP68
Other sensors, body flow meters and split converters: IP65
Spacing (separate type): The converter distance sensor generally does not exceed 100m
Second, gas turbine flowmeter measurement characteristics
1. Gas turbine flowmeter is one of the main types of flowmeters. Its working principle is: the gas entering the gas turbine flowmeter is accelerated by the fluid, and the accelerated gas acts on the blades of the turbine through the inlet passage. The impulse causes the turbine of the flow meter to rotate in the flow path, and within a certain flow range, the number of revolutions of the turbine is proportional to the volume of the gas flowing therethrough. By mechanical or electromagnetic induction, the number of revolutions f of the turbine can be obtained, so that the flow rate Q of the gas flowing through the flow meter can be obtained.
2. Gas turbine flowmeters are widely used in the measurement of gas flow or total gas volume, and are widely used in natural gas flow measurement at home and abroad. The theoretical research and practical experience of gas turbine flowmeters are very mature, and there are corresponding international standards. Countries also have corresponding national standards. China also adopts international standards. In Europe and the United States, gas turbine flow meters are the most important gas flow meters, and the Netherlands alone uses more than 2,600 gas turbine meters on natural gas pipelines.