Ultrasonic flowmeter uses the electronic principle to measure the flow meter. Although it has its own verification procedure, because it has the same usage management mode as the tap water industry measurement, it puts its performance requirements, measurement accuracy, verification cycle, etc. into the cold water meter. In the meantime, it has become more standardized. Since then, the use of large-caliber ultrasonic flowmeters has normative documents in the water industry.
Ultrasonic flowmeter (ultrasonic water meter) Because the measurement principle is different from electromagnetic water meter, its accuracy is different. How to choose an ultrasonic flowmeter with superior performance and precision:
The user flow range is used to determine the type, diameter, temperature, medium, etc. of the ultrasonic flowmeter, especially considering the common flow rate and the starting flow rate. The purpose is to ensure the water supply quantity, accurately measure and reduce the leakage rate. When using the water meter, the user's water meter Commonly used traffic generally cannot exceed the common flow value specified by the product.
Different flow rate to determine the diameter of the flow meter can meet the measurement range.
Different user environments.
Also consider the working pressure, the structure of the flow certificate, the form of the converter (whether bidirectional flow, current, pulse, communication interface, data storage) and the length of the connecting cable.
The performance and accuracy of the ultrasonic flowmeter are selected.
Users can learn more about the peers who are already using the ultrasonic flowmeter, to see the effect of their use, the accuracy of the measurement.
At present, industrial flow measurement generally has problems of large diameter, large flow, small diameter, small flow, high temperature liquid, and gas measurement. This is because the general flowmeter will bring manufacturing and transportation as the diameter of the measuring pipe increases. The difficulty of the above, the advantages of increased cost, increased energy loss, and inconvenient installation, ultrasonic flowmeters can be avoided. The German FLEXIM ultrasonic flowmeter measures diameters from 6mm to 6500 mm, from a few meters wide open channels, culverts to 500m wide rivers.
Liquid turbine flow meter classification and function description
Keywords: liquid turbine flowmeter, diesel liquid turbine flowmeter, pure water liquid turbine flowmeter
First, the liquid turbine flowmeter overview
The liquid turbine flowmeter is the main type of impeller flow (flow rate) meter, and the impeller type flowmeter also has an anemometer and a water meter. The TUF consists of a sensor and a conversion display. The sensor uses a multi-blade rotor to sense the average flow rate of the fluid, thereby deriving the flow or total amount. The rotational speed (or number of revolutions) of the rotor can be detected mechanically, magnetically, and photoelectrically and displayed and transmitted by the reading device.
Second, liquid turbine flowmeter classification
According to the classification of instrument functions, SDLWGY series turbine flowmeters can be divided into 2 categories.
1, turbine flow sensor / transmitter
2, intelligent integrated turbine flowmeter
Third, the liquid turbine flowmeter function description
1, turbine flow sensor / transmitter
This type of turbine flow product itself does not have a live display function, only the flow signal is transmitted far. The flow signal can be divided into pulse signal or current signal (4-20mA); the instrument is low in price, high in integration and small in size, and is especially suitable for use with computer control systems such as secondary display, PLC, DCS.
Fourth, the liquid turbine flowmeter put into operation the opening and closing sequence
1. The sequence of opening and closing of the operation
For flow sensors without a bypass line, first open the flow sensor upstream valve at a medium opening and then slowly open the downstream valve. Run at a small flow rate for a period of time (eg 10 minutes), then fully open the upstream valve, then open the large downstream valve opening to adjust to the required normal flow.
The flow sensor equipped with the bypass pipe first opens the bypass pipe valve to open the upstream valve at a medium opening degree, slowly opens the downstream valve, and closes the opening of the small bypass valve to make the instrument run for a period of time with a small flow rate. Then fully open the upstream valve, fully close the bypass valve (to ensure no leakage), and finally adjust the downstream valve opening to the required flow.
2. Activation of low temperature and high temperature fluids
The low-temperature fluid pipeline should drain the water in the pipeline before the flow, and then run for 15 minutes at a small flow rate, and then gradually increase to the normal flow. Slow down when stopping, so that the pipe temperature and ambient temperature are gradually approaching. High temperature fluid operation is similar to this.