Electromagnetic flowmeter features
Frequency programmable low frequency rectangular wave excitation improves stability of flow measurement and low power loss.
The new 16-bit ultra-low power microprocessor with FLASH memory has high integration, fast calculation speed and high calculation accuracy.
Full digital processing, strong anti-interference ability and reliable measurement.
Ultra-low EMI switching power supply, suitable for power supply voltage variation range, high efficiency, small temperature rise; good EMC performance.
Chinese and English menu operation, easy to use, easy to operate, easy to learn and understand.
High definition backlight wide temperature LCD display.
It can perform bidirectional flow measurement and bidirectional total accumulation; it has automatic range switching function, which can effectively improve the measurement accuracy of analog current and frequency output, especially suitable for occasions where the diurnal flow range changes greatly and needs to send control signals; flow measurement range Up to 1500:1.
There are three totalizers inside, which record and display the forward cumulative amount, the reverse cumulative amount and the accumulated difference integrated amount, which are convenient for fluid metering and custody transfer.
Provide isolated or non-isolated RS485/RS232C digital communication interface, and support fieldbus communication modes such as MODBUS, PROFIBUS-DP and HART.
Constant current source fluid resistance measurement can accurately measure the internal resistance of the electrode signal in the case of long-line transmission. It can be used not only to determine whether the fluid in the sensor is empty or not, but also to identify abnormal phenomena such as contamination and coverage of the electrode, and to provide cleaning for the user. Fault processing information such as electrodes.
Gas turbine flowmeter and screwing vortex flowmeter measurement difference
Keywords: gas turbine flowmeter, Tianxin gas turbine flowmeter, gas gas turbine flowmeter
The gas turbine flowmeter has good repeatability and is the preferred flow meter for trade settlement. The gas swirling vortex flowmeter is generally repeatable.
The gas turbine flowmeter can obtain very high frequency signals with strong signal resolution. The frequency of the gas swirling vortex flowmeter is very low, and the smaller the diameter, the lower the signal resolution. The measurement error is to be measured.
The gas turbine flowmeter has a wide range, generally 40:1~20:1, and has a wide measuring range, which can be suitable for occasions with large flow changes. When natural gas is delivered, the flow changes are generally large and, therefore, are particularly suitable for natural gas measurements. The latter range is medium, generally 15:1 to 10:1, and the measurement range is slightly narrower. When measuring gas, it is more suitable for occasions where the flow rate does not change much.
In terms of seismic performance, gas turbine flowmeters have better resistance to pipeline vibration. The gas swirling vortex flowmeter is susceptible to pipeline vibration and should be taken care of when using it.
Fourth, the liquid turbine flowmeter put into operation the opening and closing sequence
1. The sequence of opening and closing of the operation
For flow sensors without a bypass line, first open the flow sensor upstream valve at a medium opening and then slowly open the downstream valve. Run at a small flow rate for a period of time (eg 10 minutes), then fully open the upstream valve, then open the large downstream valve opening to adjust to the required normal flow.
The flow sensor equipped with the bypass pipe first opens the bypass pipe valve to open the upstream valve at a medium opening degree, slowly opens the downstream valve, and closes the opening of the small bypass valve to make the instrument run for a period of time with a small flow rate. Then fully open the upstream valve, fully close the bypass valve (to ensure no leakage), and finally adjust the downstream valve opening to the required flow.
2. Activation of low temperature and high temperature fluids
The low-temperature fluid pipeline should drain the water in the pipeline before the flow, and then run for 15 minutes at a small flow rate, and then gradually increase to the normal flow. Slow down when stopping, so that the pipe temperature and ambient temperature are gradually approaching. High temperature fluid operation is similar to this.